Evaluation of skin dose associated with different frequencies of bolus applications in post-mastectomy three-dimensional conformal radiotherapy
© Andic et al; licensee BioMed Central Ltd. 2009
Received: 22 December 2008
Accepted: 24 March 2009
Published: 24 March 2009
The study aimed to calculate chest-wall skin dose associated with different frequencies of bolus applications in post-mastectomy three-dimensional conformal radiotherapy (3D-CRT) and to provide detailed information in the selection of an appropriate bolus regimen in this clinical setting.
CT-Simulation scans of 22 post-mastectomy patients were used. Chest wall for clinical target volume (CTV) and a volume including 2-mm surface thickness of the chest wall for skin structures were delineated. Precise PLAN 2.11 treatment planning system (TPS) was used for 3D-CRT planning. 50 Gy in 25 fractions were prescribed using tangential fields and 6-MV photons. Six different frequencies of bolus applications (0, 5, 10, 15, 20, and 25) were administered. Cumulative dose-volume histograms were generated for each bolus regimen. The minimum, maximum and mean skin doses associated with the bolus regimens were compared. To test the accuracy of TPS dose calculations, experimental measurements were performed using EBT gafchromic films.
The mean, minimum and maximum skin doses were significantly increased with increasing days of bolus applications (p < 0.001). The minimum skin doses for 0, 5, 10, 15, 20, and 25 days of bolus applications were 73.0% ± 2.0%, 78.2% ± 2.0%, 83.3% ± 1.7%, 88.3% ± 1.6%, 92.2% ± 1.7%, and 93.8% ± 1.8%, respectively. The minimum skin dose increments between 20 and 25 (1.6% ± 1.0%), and 15 and 20 (4.0% ± 1.0%) days of bolus applications were significantly lower than the dose increments between 0 and 5 (5.2% ± 0.6%), 5 and 10 (5.1% ± 0.8%), and 10 and 15 (4.9% ± 0.8%) days of bolus applications (p < 0.001). The maximum skin doses for 0, 5, 10, 15, 20, and 25 days of bolus applications were 110.1% ± 1.1%, 110.3% ± 1.1%, 110.5% ± 1.2%, 110.8% ± 1.3%, 111.2% ± 1.5%, and 112.2% ± 1.7%, respectively. The maximum skin dose increments between 20 and 25 (1.0% ± 0.6%), and 15 and 20 (0.4% ± 0.3%) days of bolus applications were significantly higher than the dose increments between 0 and 5 (0.2% ± 0.2%), 5 and 10 (0.2% ± 0.2%), and 10 and 15 (0.2% ± 0.2%) days of bolus applications (p ≤ 0.003). The TPS overestimated the near-surface dose 10.8% at 2-mm below the skin surface.
In post-mastectomy 3D-CRT, using a 1-cm thick bolus in up to 15 of the total 25 fractions increased minimum skin doses with a tolerable increase in maximum doses.
Post-mastectomy radiotherapy improves survival and local control in patients with high risk breast cancer [1, 2]. The chest wall is the most frequent site of recurrence and delivering adequate radiation doses to the chest wall is crucial to reducing the risk of treatment failure . Keeping radiation-induced side effects as low as possible, while providing the intended dose to the chest wall remains a challenge [4, 5]. Definition of surface and superficial chest wall doses provides valuable information for avoiding near-surface recurrences and limiting severe early and late skin reactions.
Tissue-equivalent material boluses, which are thick enough to provide an adequate dose build-up in the skin and superficial chest wall, are commonly used during post-mastectomy radiotherapy. Skin dose contributions of boluses and the dose delivered to skin and subcutaneous tissue are important, especially in locally advanced breast cancer . The American Society of Clinical Oncology published treatment guidelines for post-mastectomy radiotherapy in 2001. These guidelines stated that the chest wall should be treated adequately but they did not comment on the use of boluses .
To our knowledge, the mean, minimum, and maximum skin doses associated with different durations of bolus applications have not been reported. The purpose of this prospective dosimetric study was to calculate the chest-wall skin dose associated with various frequencies of bolus applications in post-mastectomy three-dimensional conformal radiotherapy (3D-CRT) and to provide detailed information to aid in the selection of an appropriate bolus regimen in this clinical setting.
We performed CT-simulation of 22 patients immobilized with a breast-board. Each patient was positioned supine on the breast board with the ipsilateral arm abducted above the head; board angles were tailored according to the patient's anatomy. Patients were scanned with a 6 detector helical CT (CT Brilliance, Philips Medical Systems, Netherlands) with 5-mm slices from mid-neck to mid-abdomen.
Volumes of interest
To evaluate skin dose accurately, another volume including 2-mm surface thickness of the CTV was contoured (Figure 1) as skin structure.
The planning target volume (PTV) was defined by adding 5-mm to the CTV. However, the superficial contour of the PTV was outlined 3-mm deep to the skin surface since the build-up effect would cause apparent underdosage in the dose-volume histograms (DVH) and difficulties in the evaluation of the treatment plans.
The Precise PLAN®2.11 (Elekta, Crawley, UK) treatment planning system (TPS) was used for 3D-CRT planning. The TPS calculates the dose distribution of the photon beam using an irregular field algorithm based on data measures in a phantom for different depths and field sizes. The irregular field algorithm takes into account the tissue inhomogeneity and uses an integration scheme to evaluate the scatter component of the dose.
Although a uniform dose to the CTV within 95% to 107% of the prescribed dose is recommended, a variation of plus or minus 10% from the prescribed dose is widely used in clinical practice . In the present study, to accurately evaluate the dose contribution of later bolus applications, we planned that 90% to 110% of the prescribed dose to the PTV would be delivered before the bolus applications. Maximum doses higher than 110% of the prescribed doses were ignored if they encompassed a point and not a volume.
To test the accuracy of TPS near-surface dose calculations, solid plate phantom (Iba Dosimetry, Schwarzenbruck, Germany) and EBT gafchromic (International Specialty Products, Wayne, NJ, USA) films were used for both calibration and experimental measurements at a Synergy Platform 6-MV linear accelerator (Elekta, Crawley, UK).
For calibration, 4 × 4 cm2 films were irradiated at 100-cm fixed SSD (source-to-skin distance) and 5-cm depth with different doses ranging from 4.128 cGy (5 MU) to 336.1 cGy (400 MU). After 24 hours later, irradiated films were scanned using Epson, Expression 10000 XL (Seiko Epson Corporation, Japan) scanner, read with Mephysto mc2 v1.3 (PTW, Freiburg, Germany) software and optic density-dose calibration curves were obtained.
For dose measurements, 4 × 4 cm2 films were placed at the centre of the 10 × 10 cm2 field at specific depths (0, 1, 2, 3, 4, 5, 6, 8, 10, 12, 15, 20, 25 and 30-mm) and irradiated at 100-cm fixed SSD with a dose of 83.25 cGy (100 MU).
We analyzed the dosimetric data of skin structure for all treatment regimens. The mean, minimum and maximum doses to skin in all bolus regimens were compared by the Friedman test and Wilcoxon analysis using Statistical Package for Social Sciences (SPSS), version 16.0. P-values of 0.05 or less were considered statistically significant. Values are expressed as mean (range) ± standard deviation (SD) and percent of prescribed dose.
The mean, minimum and maximum PTV doses before the bolus applications were 101.8% (100.2–103.2%) ± 0.9%, 91.2% (90.0–94.5%) ± 1.2% and 109.4% (105.0–110.6%) ± 1.3%, respectively.
Mean values of mean, minimum, and maximum skin structure doses according to bolus frequencies
Mean ± SD*
Minimum ± SD*
Maximum ± SD*
100.0 ± 1.1
73.0 ± 2.0
110.1 ± 1.1
100.6 ± 1.1
78.2 ± 2.0
110.3 ± 1.1
101.3 ± 1.1
83.3 ± 1.7
110.5 ± 1.2
101.9 ± 1.1
88.3 ± 1.6
110.8 ± 1.3
102.6 ± 1.1
92.2 ± 1.7
111.2 ± 1.5
103.2 ± 1.1
93.8 ± 1.8
112.2 ± 1.7
Bolus use in all fractions provided a 20.8% ± 2.8% minimum skin dose increment. The minimum skin dose increments between 20 and 25 (1.6% ± 1.0%), and 15 and 20 (4.0% ± 1.0%) days of bolus applications were significantly lower than the dose increments between 0 and 5 (5.2% ± 0.6%), 5 and 10 (5.1% ± 0.8%), and 10 and 15 (4.9% ± 0.8%) days of bolus applications (p < 0.001). Furthermore, the minimum skin dose increment between 20 and 25 (1.6% ± 1.0%) days of bolus application was lower than the dose increment between 15 and 20 (4.0% ± 1.0%) days of bolus application (p < 0.001).
Bolus use in all fractions resulted in a 2.0% ± 1.2% maximum skin dose increment. The maximum skin dose increments between 20 and 25 (1.0% ± 0.6%), and 15 and 20 (0.4% ± 0.3%) days of bolus applications were significantly higher than the dose increments between 0 and 5 (0.2% ± 0.2%), 5 and 10 (0.2% ± 0.2%), and 10 and 15 (0.2% ± 0.2%) days of bolus applications (p ≤ 0.003). Furthermore, the maximum skin dose increment between 20 and 25 (1.0% ± 0.6%) days of bolus application was higher than the dose increment between 15 and 20 (0.4% ± 0.3%) days of bolus application (p < 0.001).
The dose increase of the mean values between all bolus frequencies was similar (p= 0.965).
Measurements using EBT gafchromic film revealed that Precise PLAN®2.11 TPS overestimated near-surface dose 10.8% at 2-mm below the skin surface.
Bolus thickness required to enhance surface dose is optimized according to surface and build-up region dosimetry. In the present study, a 1-cm bolus was used to increase skin doses. This thickness was chosen because 6-MV photon energy with a 1.5-cm maximal depth was used for tangential fields.
The skin dose contributions of 1-cm bolus material during whole or a part of treatment duration were calculated in this study. The results showed a trend of increasing minimum skin dose when the days of bolus application were increased. The minimum skin dose increments were expected to be linear among the bolus durations. However, the minimum skin dose increments between 20 and 25 (1.6% ± 1.0%), and 15 and 20 (4.0% ± 1.0%) days of bolus applications were significantly lower than the dose increments between 0 and 5 (5.2% ± 0.6%), 5 and 10 (5.1% ± 0.8%), and 10 and 15 (4.9% ± 0.8%) days of bolus applications while the maximum skin dose increments were significantly higher. TPS dose calculation algorithm and treatment related factors such as delivery technique, field size and angle of beam incidence are supposed to be associated with these non-linear dose increments. Therefore, our results need to be clarified in further dosimetric studies using different TPS, techniques, beam energies, and bolus thicknesses.
Determining the necessary frequency of bolus treatments is critically important in post-mastectomy radiotherapy, since it influences the irradiated volume as well as the skin doses. Although the literature contains several recommendations for radiotherapy planning techniques, there are few recommendations regarding bolus use [4, 5, 9–11]. The optimal duration and the optimal thickness of the bolus material still remain uncertain and change centre to centre [7, 12]. Wide regional variations in the use of boluses were reported by Vu et al. in an international survey of radiation oncologists and their opinions on the indications for boluses in post-mastectomy radiotherapy .
Determining the difference between the calculated and measured surface dose is useful when evaluating and comparing patient plans and also when optimizing the use of boluses. Many factors affect the magnitude of the surface dose, such as the delivery technique, field size, angle of beam incidence, air gap and the use of bolus material and beam modifiers [13–15]. Calculation of skin doses is difficult in most TPSs due to their inability to account for all the factors that contribute to the surface dose. However, the Monte Carlo TPSs and, to a lesser extent, the modern true 3D algorithms are able to calculate skin doses [16–18]. Doses calculated with different TPSs have been reported to underestimate and overestimate measured skin doses [15, 19–23]. Measured skin doses also may differ according to the dosimetry used . In the present study, EBT gafchromic films were used to test the accuracy of TPS and experimental measurements were performed at 10 × 10 cm2 field at 100-cm fixed SSD since same tissue equivalent field as in a post-mastectomy patient could not be created to measure depth doses. The measurements revealed that Precise PLAN®2.11 TPS overestimated the near surface dose comparable to the literature [15, 19–22]. However, the goal of the present study was to reveal the trend in proportional skin doses with various frequencies of bolus applications, using the same TPS to calculate doses to the same skin structures.
The thickness of the epidermis varies between 0.05–1.5 mm, depending on the anatomic location. The International Commission on Radiological Protection and the International Commission on Radiation Units and Measurements recommends a depth of 0.07-mm, corresponding to the epidermal and dermal layers, for practical skin dose assessments [24, 25]. Measuring the dose at that depth is very difficult. Therefore, in the present study, skin structure was defined as 2-mm surface thickness of the CTV. Court et al. also used a 2-mm thick skin structure in their investigation of the accuracy of skin dose calculations on a semi-cylindrical model of a neck or breast .
The superficial PTV contour is usually outlined 5-mm deep to the skin surface to avoid apparent under-dosage in the DVH due to build-up effects [4, 26]. Although this is reasonable in breast conserving surgery, it may result in wrong dose-volume information in post-mastectomy radiotherapy, particularly in locally advanced breast cancer when the skin is close to or included in the target volume. Therefore, we believe that delineating a skin structure in addition to the CTV and PTV would provide important information in post-mastectomy treatment planning. Furthermore, surface dose measurements for the comparison of calculated and measured skin doses would also help to define accurate skin dose deficit.
In post-mastectomy 3D-CRT, using a 1-cm thick bolus in 5, 10, and 15 of the total 25 fractions increased minimum skin doses with a tolerable increase in maximum doses. Hence, up to 15 days of bolus applications appear to be the optimal bolus regimens. However, while deciding duration of bolus application, the difference between calculated and measured skin doses should also be considered, besides the calculated skin dose deficit in the TPS.
three-dimensional conformal radiotherapy
clinical target volume
treatment planning system
planning target volume
Statistical Package for Social Sciences
- Overgaard M, Hansen PS, Overgaard J, Rose C, Andersson M, Bach F, Kjaer M, Gadeberg CC, Mouridsen HT, Jensen MB, Zedeler K: Postoperative radiotherapy in high-risk premenopausal women with breast cancer who receive adjuvant chemotherapy. Danish Breast Cancer Cooperative Group 82b Trial. N Engl J Med. 1997, 337: 949-955.View ArticleGoogle Scholar
- Whelan TJ, Julian J, Wright J, Jadad AR, Levine ML: Does locoregional radiation therapy improve survival in breast cancer? A meta-analysis. J Clin Oncol. 2000, 18: 1220-1229.Google Scholar
- Taylor ME, Perez CA, Mortimer JE, Levitt SH, Ieumwananonthachai N, Wahab SH: Breast: Locally Advanced (T3 and T4) and Recurrent Tumors. Principles and Practices of Radiation Oncology. Edited by: Perez CA, Brady LW, Halperin EC, Schmidth-Ullrich RK. 2004, Philadelphia, USA: Lippincott Williams and Wilkins Press, 1502-1553. 4Google Scholar
- Cho BC, Hurkmans CW, Damen EM, Zijp LJ, Mijnheer BJ: Intensity modulated versus non-intensity modulated radiotherapy in the treatment of the left breast and upper internal mammary lymph node chain: a comparative planning study. Radiother Oncol. 2002, 62: 127-136.View ArticleGoogle Scholar
- Sonnik D, Selvaraj RN, Faul C, Gerszten K, Heron DE, King GC: Treatment techniques for 3D conformal radiation to breast and chest wall including the internal mammary chain. Med Dosim. 2007, 32: 7-12.View ArticleGoogle Scholar
- Thoms WW, McNeese MD, Fletcher GH, Buzdar AU, Singletary SE, Oswald MJ: Multimodal treatment for inflammatory breast cancer. Int J Radiat Oncol Biol Phys. 1989, 17: 739-745.View ArticleGoogle Scholar
- Recht A, Edge SB, Solin LJ, Robinson DS, Estabrook A, Fine RE, Fleming GF, Formenti S, Hudis C, Kirshner JJ, Krause DA, Kuske RR, Langer AS, Sledge GW, Whelan TJ, Pfister DG, American Society of Clinical Oncology: Postmastectomy radiotherapy: clinical practice guidelines of the American Society of Clinical Oncology. J Clin Oncol. 2001, 19: 1539-1569.Google Scholar
- ICRU 50: Prescribing, recording, and reporting photon beam therapy. 1993, Bethesda, MD: International Commission on Radiation Units and Measurements PressGoogle Scholar
- Thomsen MS, Berg M, Nielsen HM, Pedersen AN, Overgaard M, Ewertz M, Block T, Brodersen HJ, Caldera C, Jakobsen E, Kamby C, Kjaer-Kristoffersen F, Klitgaard D, Nielsen MM, Stenbygaard L, Zimmermann SJ, Grau C, Danish Breast Cancer Cooperative Group: Post-mastectomy radiotherapy in Denmark: from 2D to 3D treatment planning guidelines of The Danish Breast Cancer Cooperative Group. Acta Oncol. 2008, 47: 654-661.View ArticleGoogle Scholar
- Pierce LJ, Butler JB, Martel MK, Normolle DP, Koelling T, Marsh RB, Lichter AS, Fraass BA: Postmastectomy radiotherapy of the chest wall: dosimetric comparison of common techniques. Int J Radiat Oncol Biol Phys. 2002, 52: 1220-1230.View ArticleGoogle Scholar
- Ung YC, Sixel KE, Bell C: The influence of patient geometry on the selection of chest wall irradiation techniques in post-mastectomy breast cancer patients. Radiother Oncol. 2000, 57: 69-77.View ArticleGoogle Scholar
- Vu TT, Pignol JP, Rakovitch E, Spayne J, Paszat L: Variability in radiation oncologists' opinion on the indication of a bolus in post-mastectomy radiotherapy: an international survey. Clin Oncol (R Coll Radiol). 2007, 19: 115-119. 10.1016/j.clon.2006.10.004.View ArticleGoogle Scholar
- Quach KY, Morales J, Butson MJ, Rosenfeld AB, Metcalfe PE: Measurement of radiotherapy X-ray skin dose on a chest wall phantom. Med Phys. 2000, 27: 1676-1680.View ArticleGoogle Scholar
- Hsu SH, Roberson PL, Chen Y, Marsh RB, Pierce LJ, Moran JM: Assessment of skin dose for breast chest wall radiotherapy as a function of bolus material. Phys Med Biol. 2008, 53: 2593-2606.View ArticleGoogle Scholar
- Court LE, Tishler R, Xiang H, Allen AM, Makrigiorgos M, Chin L: Experimental evaluation of the accuracy of skin dose calculation for a commercial treatment planning system. J Appl Clin Med Phys. 2008, 9: 2792-2799. [http://www.jacmp.org/index.php/jacmp]Google Scholar
- Mackie RT, Liu HH, McCullough EC: Treatment Planning Algorithms: Model-based Photon Dose Calculations. Treatment Planning in Radiation Oncology. Edited by: Khan FM. 2007, USA: Lippincott Williams and Wilkins Press, 63-77. 2Google Scholar
- Oelfke U, Scholz C: Dose Calculation Algorithms. New Technologies in Radiation Oncology. Edited by: Schlegel W, Bortfeld T, Grosu A-L. 2006, Berlin, Germany: Springer-Verlag Press, 187-196. 1Google Scholar
- Fippel M: Monte Carlo Dose Calculation for Treatment Planning. New Technologies in Radiation Oncology. Edited by: Schlegel W, Bortfeld T, Grosu AL. 2006, Berlin, Germany: Springer-Verlag Press, 197-206. 1View ArticleGoogle Scholar
- Chung H, Jin H, Dempsey JF, Liu C, Palta J, Suh TS, Kim S: Evaluation of surface and build-up region dose for intensity-modulated radiation therapy in head and neck cancer. Med Phys. 2005, 32: 2682-2689.View ArticleGoogle Scholar
- Ramsey CR, Seibert RM, Robison B, Mitchell M: Helical tomotherapy superficial dose measurements. Med Phys. 2007, 34: 3286-3293.View ArticleGoogle Scholar
- Cheek D, Gibbons JP, Rosen II, Hogstrom KR: Accuracy of TomoTherapy treatments for superficial target volumes. Med Phys. 2008, 35: 3565-73.View ArticleGoogle Scholar
- Roland TF, Stathakis S, Ramer R, Papanikolaou N: Measurement and comparison of skin dose for prostate and head-and-neck patients treated on various IMRT delivery systems. Appl Radiat Isot. 2008, 66: 1844-1849.View ArticleGoogle Scholar
- Mutic S, Low DA: Superficial doses from serial tomotherapy delivery. Med Phys. 2000, 27: 163-165.View ArticleGoogle Scholar
- ICRP 2007: The 2007 Recommendations of the International Commission on Radiological Protection: adopted by the Commission in March 2007. 2007, Essen, Germany: Elsevier PressGoogle Scholar
- ICRU 39: Determination of dose equivalents resulting from external radiation sources. 1985, Bethesda, MD: International Commission on Radiation Units and Measurements PressGoogle Scholar
- Landau D, Adams EJ, Webb S, Ross G: Cardiac avoidance in breast radiotherapy: a comparison of simple shielding techniques with intensity-modulated radiotherapy. Radiother Oncol. 2001, 60: 247-255.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.