Monitoring of people and workers exposure to the electric, magnetic and electromagnetic fields in an Italian national cancer Institute

Background The paper reports the electric, magnetic and electromagnetic fields (emf) measurements carried out in the Regina Elena National Cancer Institute (NCI). Several devices, used in diagnostics and in medical cures, can represent sources of emf for the workers and for the public subjected to the treatments. The aim is to evaluate their exposition, in order to assess the compliance with the law. Methods The investigations have been carried out in the departments of: intensive care, physiotherapy, MR presstherapy and in the surgical rooms. The measurements have been performed using broad band probes in the frequency ranges 5 Hz÷30 kHz and 100 kHz-3 GHz. Results The variability of the magnetic induction (B(μT)) levels is between 0,05 μT and 80 μT. The statistical distribution shows that most of the measurements are in the range 0,05<B = 0,5 μT and the 89% of the B(μT) levels are within the 3 μT. Conclusion The measurement of the emf levels in the NCI is recommended because of the presence of the oncological patients; their long stay near the equipments and their day-long exposure represent additional risk factors for which a prudent avoidance strategy have to de adopted.


Background
In the last years, the increase in the number of systems for telecommunications and for the electric energy transport, has produced a huge debate both in the scientific community and among the public on the potential risks on health due to long time exposition to the electric field (ef), magnetic field (mf) and electromagnetic field (emf) in the working and living environments. The effects produced by the interaction of these physical agents with the biological systems depend on frequency, amplitude of the fields and on the time of the exposition. The effects can be classified in thermal and non-thermal [1]. The former are scientifically well-know and consist in heat production in the biological system; the exposure limits recommended by the national and international legislation protect from this injury. The existence of the non-thermal effects represents an open issue; they concern the evidence of biological changes without an effective increase of temperature. The scientific debate is centred on the issue if a long exposition to ef-mf-emf levels lower than the limits could cause harmful effects on health. Many national and international research projects are involved in this investigation by in vivo, in vitro and epidemiological studies. [2][3][4][5][6][7]. Until now, conclusive results are still not available.
In Italy, the issue of people exposition to the ef-mf-emf is regulated by two decrees promulgated in 2003, one for fields produced by sources with frequencies between 100 kHz and 300 GHz [8], the other for power lines at the frequency of 50 Hz [9]. Their distinctive feature is to establish people long term exposure levels to ef-mf-emf lower than the limits recommended by the international standards fixed by the International Commission on non-ionizing radiation protection (ICNIRP) [10]. For sources not referable to power lines in the frequency range 0 Hz-100 kHz, the whole of the restrictions fixed by the European Council Recommendation of 12 July 1999 [11] has to be considered. About the workers, the directive 2004/40/EC [12], recently incorporated into the Italian legislation system with the decree n. 257/2007 [13], has to be applied.
The main aim of this paper is to report the ef-mf-emf measurements carried out in the Regina Elena National Cancer Institute (NCI). Several electric and electronic devices, used in diagnostics and in medical cures, have been investigated, which can represent potential sources of ef-mf-emf for the health staff, for the public subjected to the treatments and also for their accompanists. The main peculiarity of this study is then to verify the exposition to these physical agents for a class of particularly vulnerable patients, as the oncological ones, in order to assess the compliance with the law. The long stay of these patients near the equipments and their day-long exposure can represent additional risk factors for which a prudent avoidance strategy has to be adopted.

Methods
The investigations have been carried out in the intensive care division, the physiotherapy department, the surgical rooms, the Magnetic Resonance (MR) department and the press-therapy room. For each division, the investigated equipments are reported in Table 1. The study has been extended also to a wireless equipment for internal communications and to external sources.
Initially, the technical characteristics of each equipment have been collected. Moreover, their operating modalities during diagnostic and therapy sessions, and their localizations in each area have been acquired, in order to detect potential interactions.
The measurements have been performed using a broad band probe (EMR 300 Wandel & Goltermann) providing the total electric field for the frequency range 100 kHz -3 GHz. According with the Italian decrees [8,9], the length of time of each measurement was six minutes. For the frequencies range 5 Hz÷30 kHz, the EFA-3 Wandel & Goltermann, equipped with an external magnetic probe, has been used. All the measurements have been carried out in the operating clinical conditions.
The check on the external sources has been executed using the narrow band device SRM3000 Narda, equipped with  1) and specific equipments, conveniently placed in one box (Fig. 2). B(μT) levels are generally lower than 1 μT, with few exceptions, near the tapis roulant (4,5 μT), and near the press-therapy equipment, where a maximum B(μT) level of 1,6 μT has been registered. Effective electric field (E eff (V/m)) levels lower than 2,3 V/m have been registered near the equipment for ultrasound therapy. Further measurements, performed near the head of the Laser Power Diode, have provided E eff (V/m) measurements in the range 1,0 -3,0 V/ m.

(b) Press-therapy room
The rehabilitation area consists of: the reception, the press-therapy room and a medical office. In particular, in the press-therapy room, two equipments for the arms Lymphatic Drainage are placed. The duration of this therapeutic treatment varies from 2 hours up to 4 hours and is generally performed for four patients at the same time.
The B(μT) levels, measured in the points reported in Fig.  3, are reported in Table 4.
During the treatments, the B(μT) levels detected near the patients are lower than 1 μT. Around the equipments, in positions where only the staff could eventually stay, B(μT) levels between 1,1 and 5,0 μT have been registered.
Higher levels (up to 80 μT) have been obtained near the air compressors and the electronics of the equipments, in the positions reported in Table 4. Additional measurements, performed at a height of about 10 cm from the floor, have provided B(μT) levels lower than 1,4 μT. Further investigations, carried out in the contiguous rooms, during the equipments standard working conditions, provided B(μT) values between 0,1 and 0,2 μT.

(c) Intensive care division
The intensive care division consists of an open space, where three patients' beds and two monitoring workplaces are located, and three rooms. In Table 5, the results of the B(μT) measurements in the open space (Fig. 4) near the sources listed in Table 1, are reported; the magnetic induction measurements, carried out at a height of 1,0 m and 1,9 m from the floor, near the electric panel and near the intensive care staff workplaces, are lower than 1,9 μT. Magnetic field background measurements in the open space provided B(μT) levels equal to 70 nT. The high frequencies E eff (V/m) values, measured near a patient's bed, resulted lower than 1,0 V/m. In Table 5, the results of the B(μT) measurements around a bed (Fig. 5) are reported. The B(μT) levels close to the patients are lower than 1 μT, while levels up to 7,0 μT have been detected at 1 m from the bed, at a distance of few centimetres from the volumetric and syringe infusion pumps. Higher B(μT) levels (equal to 14 μT) have been measured at the bottom of the bed near the engine of the anti-bedsore mattress.

(e) MR room and other internal sources
High frequencies E eff (V/m) measurements near the MR and into the gate room during the sequences activation, showed levels lower than 1,0 V/m.

(f) External sources
Further investigations have been carried out outside the NCI. In a radius of about 300 m a radiobase station for cellular communications is placed on the roof of a building. In the distance of a few km from the NCI, a tower with several broadcasting systems for radio, TV, and cellular radiobase stations is located. Investigations with the broad and the narrow band tools in the external area of the NCI show that these sources don't give a significant contribution to the electromagnetic field background in the area. Additional investigations inside the NCI, in particular in the physiotherapy division, allowed to detect a Points of measurements near the tapis roulant (detected fre-quency: 50 Hz) Figure 1 Points of measurements near the tapis roulant (detected frequency: 50 Hz). Specific equipments in a physiotherapy box Figure 2 Specific equipments in a physiotherapy box.
weak electric field at the frequency of 1,9 GHz produced by a radiobase station for the cellular networks (Fig. 6).
In Italy, people exposure to the ef-mf-emf is defined by two decrees [8,9] in terms of three levels: (i) limits, levels that haven't to be exceeded in any circumstances for the protection from short terms effects; (ii) attention levels that have not to be exceeded into gambling areas, schools and buildings where people stay for more than 4 hours, as a precaution from the potential risks on health due to long time exposition; (iii) quality targets, levels to achieve in a long period in order to minimize people exposure. The respective levels for the protection of people exposure at high and low frequencies are reported in Tables 7 and 8.
The European recommendation [11] establishes basic restrictions and reference levels for limiting exposure.  Table 9.
The directive 2004/40/EC on the minimum health and safety requirements regarding the exposure of workers to the risks arising from electromagnetic fields, has been issued in the European Official Journal on May 25, 2004, together with its recent application in the Italian regulation [13]. It introduces measures protecting workers from the risks associated with emf, owing to their effects on the health and safety. The long-term effects, including possible carcinogenic effects due to exposure to time-varying efmf-emf for which there is no conclusive scientific evidence establishing a causal relationship, are not addressed. Expo-Results of the magnetic induction measurements near the press-therapy equipments Figure 3 Results of the magnetic induction measurements near the press-therapy equipments.  sure limit values and action values have been defined. The former are based directly on established health effects and biological considerations and compliance with these limits ensures that workers exposed to emf are protected against all known adverse health effects. The action values refer to the magnitude of directly measurable parameters, provided in terms of E, H, B and S; the observance with these values ensure compliance with the relevant exposure limit. Action values are obtained from the exposure limit values according to the rationale used by the International Commission on Non-ionising Radiation Protection (ICNIRP) in its guidelines on limiting exposure to non-ionising radiation [13].
For the health staff in the NCI, the exposure levels are in agreement with the recommendations fixed by the international standards. However, for the health staff, the more restrictive Italian people long term exposure levels have been established as investigation levels. The results of the monitoring in the working areas frequented by patients and accompanies, near the equipments reported in Table 1 show a variability of the B(μT) measurements between 0,01 and 80 μT. The global statistical distribution of the B(μT) levels shows that most of the measurements are in the range 0,05<B = 0,5 μT, with 25% of the levels registered in the surgical room, 9% in the press-therapy room and 15% in the physiotherapy division. The 89% of the B(μT) levels measured near electric and electronic devices used in diagnostics and in medical cures, in positions normally occupied by the patients during the treatments are within 3 μT, while 8% are within 10 μT. Only 3% of the measurements is higher than this level but Measurements near a patient's bed in the intensive care divi-sion Figure 5 Measurements near a patient's bed in the intensive care division.
Results of the spectrum analysis carried out in the physiotherapy division Figure 6 Results of the spectrum analysis carried out in the physiotherapy division.
within the restrictive limits imposed by the Italian law to regulate the exposure of the public to the mf. The last levels have been measured typically near the engine of the tools and near the electric power of the instrumentations in areas where the presence of the population is not scheduled. In order to minimize or reduce exposures, it is then sufficient to set the area around the devices, with the indications to the staff not to stay close to the devices in their operative conditions.
The exposure conditions of the patients for the extremely low frequencies and the emf in the intensive care are in accordance with the measurements carried out by Petrucci [14]. The results show that the type of equipments, their number and also their orientation respect to the patient is important to minimize mf levels; in general, a collection of equipments should be avoided and a safety distance have to be adopted to avoid unnecessary exposures.
About the possible electromagnetic interference (EMI) with electronic equipments by radio waves coming from outside the hospital, we find E eff (V/m) sensibly lower than the values registered by Hanada et others [15]. However, measurements of the electromagnetic environment should be performed by each hospital specially in urban areas [16] where some sources could potentially induce strong electric field intensity causing malfunctions.