High performance of targeted next generation sequencing on variance detection in clinical tumor specimens in comparison with current conventional methods

Background Next generation sequencing (NGS) is being increasingly applied for assisting cancer molecular diagnosis. However, it is still needed to validate NGS accuracy on detection of DNA alternations based on a large number of clinical samples, especially for DNA rearrangements and copy number variations (CNVs). This study is to set up basic parameters of targeted NGS for clinical diagnosis and to understand advantage of targeted NGS in comparison with the conventional methods of molecular diagnosis. Methods Genomic DNA from 1000 Genomes Project and DNA from cancer cell lines have been used to establish the basic parameters for targeted NGS. The following confirmation was conducted by clinical samples. The multiple variants tested by amplification-refractory mutation system (ARMS), fluorescence in situ hybridization (FISH) and immunohistochemistry (IHC) were evaluated by targeted NGS to determine the sensitivity. Furthermore, the multiple variants detected by targeted NGS were confirmed by current conventional methods to elucidate the specificity. Results At sequencing depth of 500×, the maximal sensitivities on detecting single nucletic variances (SNVs) and small insertions/deletions (Indels) can reach 99% and 98.7% respectively, and in 20% of cancer cells, CNV detection can reach to the maximal level. The following confirmation of the sensitivity and specificity was conducted by a large cohort of clinical samples. For SNV and indel detection in clinical samples, targeted NGS can identify all hotspot mutations with 100% sensitivity and specificity. On ALK fusion detection, about 86% IHC-identified cases could be identified by targeted NGS and all ALK fusion detected by targeted NGS were confirmed by IHC. For HER2-amplification, 14 HER2-amplification cases identified by target NGS were all confirmed by FISH and about 93.3% of Her-2 IHC (3+) cases were identified by targeted NGS. Finally, the targeted NGS platform developed here has accurately detected EGFR hotspot mutations in 215 NSCLC patients. Conclusions DNA from cancer cell lines is better than standard DNA as a reference to establish basic parameters for targeted NGS. Comparison of the conventional methods using a large cohort of patient samples confirmed the high preformance of targeted NGS on detecting DNA alterations. Electronic supplementary material The online version of this article (10.1186/s13046-017-0591-4) contains supplementary material, which is available to authorized users.

(Continued from previous page) Conclusions: DNA from cancer cell lines is better than standard DNA as a reference to establish basic parameters for targeted NGS. Comparison of the conventional methods using a large cohort of patient samples confirmed the high preformance of targeted NGS on detecting DNA alterations.
Keywords: Targeted next generation sequencing, Amplification-refractory mutation system, Fluorescence in situ hybridization, Immunohistochemistry, Clinical tumor samples

Background
Cancer is considered to be caused by both inherited and acquired genomic alterations, which leads to uncontrolled cell growth. Over the past 20 years, new-drug development has focused on the known oncogenic drivers and heralded an era of targeted therapies. Compared to traditional cytotoxic chemotherapy, targeted therapies are safer, more efficacious and less side effects. [1] For example, it has been reported the efficacy of targeted drugs, herceptin to patients with ERBB2-amplified breast cancer and gefitinib and erlotinib to patients with mutated EGFR lung cancer, is better. So far, hundreds of targeted drugs have been developed or under development, targeting the corresponding genomic alterations, including site mutations, insertions and deletions (Indels), copy number variants (CNVs) and DNA rearrangements. Identification of these alterations in cancer patients is the first step to provides the targets for therapy. The cancer biology is complex. For instance, the patients with EGFR mutated lung cancer are benefit from erlotinib and gelfitinib, but if EFGR harboring T790 M mutation, then resistant to these drugs. [2,3] These suggest it is important to fully understand the DNA alterations in cancer patients.
Currently, the conventional technologies for identifying the genomic alterations in patients include the amplification-refractory mutation system (ARMS), fluorescence in situ hybridization (FISH) and immunohistochemistry (IHC). All of these methods have both advantages and limitations in application. These methods are well installed and highly reliable, but they have a common shortness: each genomic alteration is analyzed in a specific assay. The sensitivity of ARMS to detect site mutation from genomic DNA can reache to 0.10%, [4] but the technology is only used to detect the known base substitutions or Indels. ARMS can also be used for gene fusion detection at mRNA level, but the good quality RNA could be limited from formalin fixed, paraffin-embedded (FFPE) tissues. FISH is used to detect DNA rearrangements and amplifications at genomic level. This method is relatively rapid, well-standardized, rather expensive method. However, FISH can not distinguish fusion variants. IHC mainly detected the changes of gene expression at the protein level, which usually resulted in by gene amplification or DNA rearrangement.
Next generation sequencing (NGS) is the most powerful tool to accurately detect most gene alterations on a massive scale, allowing interrogation of all genes or selected genes in a single assay. This technology requires low amounts of DNA, and has high sensitivity and specificity. [5] Moreover, cancers are frequently caused by alterations on multiple genes, which collaborate to promote tumor development. [6] A combination of drugs targeting the multiple alterations may be an approach to achieve the best therapeutic efficacy. [7] The conventional methods are impossible to massively screen cancer-related genes in a single assay. Therefore, NGS has been increasingly used in clinical diagnosis. However, we need throughly validate the sensitivity and accuracy of NGS to detect multiple types of DNA alterations from a large number of clinical specimens, which have been confirmed by the current clinical methods.
In this study, we developed and validated a panel for targeted NGS which is able to detect multiple types of genomic alterations in 365 genes commonly associated with cancers. A number of clinical samples were collected, including 131 specimens for base substitutions and Indels, 18 for ALK fusions, 86 for HER2 amplifications. These 235 clinical tissues were used to throughly compare the results from targeted NGS and conventional methods, including ARMS, FISH or IHC, and explore the cause of disconcordance. Finally, the capability of targeted NGS on detecting multiple types of genomic alterations was performed in 215 non-small-cell lung cancer (NSCLC) samples. The results may guide the clinician to select the right method for the diagnosis based on the characteristics of each method and clinical needs.

Standard DNA, cell lines and clinical tumor specimens Clinical cancer specimens
This study was approved by Ethics Committee of Zhejiang Province Cancer Hospital,China. We have enrolled 235 formalin fixed, paraffin-embedded (FFPE) tumour specimens from lung adenocarcinoma, colon and other types of cancers in this study (Additional file 1: Table S1). The specimens have been reviewed by a pathologist to make sure the content of cancer cells is > = 20%.

Standard DNA
For base substitution validation, purified DNA from 15 lymphoblastoid cell lines from the 1000 Genomes Project were purchased from the Coriell Institute (Additional file 2: Table S2).

Cell lines
For indel validation, 28 immortalized tumor cell lines (Additional file 3: Table S3) were purchased from American Type Culture Collection (ATCC) (http://www.atcc.org/). For copy number validation, HCC1143 matched tumor and normal cell lines were purchased from ATCC as either cell pellets or DNA.
Targeted NGS Pathological examinations of the clinical tumor specimens 4-μm paraffin sections out of the clinical tumor specimens were stained with hematoxylin and eosin (HE) for pathological review to determine that a sample has a volume of ≥1 mm 3 and ≥20% tumor cells. If the percentage of tumor cells was ≤20%, a macro-dissection was used for enrichement of tumor cells.

DNA extraction
Paraffin in Formalin Fixed Paraffin Embedded (FFPE) sections and cores was removed by xylenes, followed by ethanol washing. Tissues were digested by proteinase K at 56°C overnight and incubated at 90°C for 5 min to reverse DNA crosslink. Genomic DNA was then extracted with QIAamp DNA FFPE Tissue Kit (Qiagen) and quantified by PicoGreen fluorescence assay (Invitrogen).

Construction of sequencing libraries
50-200 ng of DNAs were fragmented to around~200 bp by sonication (Covaris), and constructed into the libraries with KAPA Hyper Prep Kit (Kapa Biosystems). [8] Capture of the targeted DNAs and sequencing The baits, a pool of 16,198 individually synthesized 5′biotinylated 120 bp DNA oligonucleotides (Integrated DNA Technology), cover 4557 exons of 365 cancerrelated genes, 47 introns of 25 genes frequently rearranged in cancer (Additional file 4: Table S4). Intronic baits were filtered for repetitive elements as defined by the UCSC Genome RepeatMasker track. [9] The targeted regions were captured with the baits as described previously. [10] Briefly, a pool of indexed sequencing libraries, total 1000 ng, was lyophilized and resuspended in water, heated to denature and kept at 68°C. Then the bait, Cot, salmon sperm and adaptor-specific blocker DNA were added in the pool. After incubation, the library-bait duplexes were captured with Dynabeads M270 Streptavidin (Invitrogen) and off-target parts in the libraries were washed off by SSC. The PCR master mix was added to directly amplify the captured libraries, and followed by purification with 1.8 × SPRI, quantification by Qubit 3.0 (Life Technologies) and determination of the DNA size on LabChip GX (Caliper). Libraries were adjusted to 1.05 nM and seqeuneced in next generation sequencing platform illumina Nextseq 500.

Analysis of DNA alterations
Sequence data processing Sequence data were mapped to the human genome (hg19) using BWA aligner v0.7.12. PCR duplicate read removal and sequence metric collection were done using Picard 1.130 (https://github.com/broadinstitute/picard/ releases/tag/1.130) and Samtools 0.1.19. Variant calling was done only in the targeted genomic regions.

Base substitution analysis
We used a Bayesian methodology, which allows detection of novel somatic mutations at low mutation allele frequency (MAF) and increase sensitivity for calling mutations at hotspots through the incorporation of tissuespecific prior expectations. The total reads in the variant position could not be less than 30, and the maximum variant frequency of normal controls was 0.03. Final calls were cut off at MAF > 1% (MAF > 0.5% at hotspots) after filtering for strand bias.

Indel analysis
To detect Indels, de novo local assembly in each targeted exon was performed using the de Bruijn approach. Pindel version 0.2.5a7 (https://github.com/genome/pindel/releases/tag/v0.2.5a7) was used to detect indels in this research. Filtering of Indel candidates was carried out as described for base substitutions above (strand bias >0.9 or <0.1, MAF threshold <1% while MAF < 0.5% at hotspots), with an empirically increased requirement at repeats.

CNA analysis
A statistically rigorous and computationally efficient algorithm called BIC-seq was used for detecting CNVs. In this algorithm, a poisson or other parametric models are not assumed on the read distribution as is done in other currently available methods, and it is thus more robust to outliers and datasets that cannot be well approximated with a parametric model. It is also fast and able to handle high-coverage genomes effectively. Furthermore, the statistical framework behind BIC-seq can be extended to the problem of identifying recurrent CNVs in multiple cancer genomes. We obtained a log-ratio profile of the sample by normalizing the sequence coverage obtained at all exons against a process-matched normal control. This profile was corrected for GC-bias.

DNA arrangement analysis
Genomic rearrangements were identified by analyzing the clipped reads which can be extracted by the tag information of bam files mapped by bwa software. Then candidate reads which are discordant or with the same direction are performed to be filtered. Read pairs for which reads mapped to separate chromosomes, or at a distance of over 2 kb are kept for fusion detection in probe level. Output rearrangements contain translocation, inversion, long deletion, etc.

Amplification-refractory mutation system (ARMS) PCR
Mutational analyses of the EGFR, KRAS, NRAS and BRAF in 34 FFPE samples were carried out by ADx-ARMS Test Kits (Xiamen AmoyDx Biomedical Technology Co., Ltd.) in Zhejiang Province Cancer Hospital. Mutational analyses of the EGFR and KRAS in 97 FFPE samples from lung adenocarcinomas and colorectal cancers were carried out according to the ARMS method using Human EGFR Gene Twenty-nine Mutations Detection Kit and Human KRAS Gene Seven Mutations Detection Kit (PCR fluorescence probe method) (Wuhan YouZhiYou Medical Technology Co., Ltd.). Some EGFR mutations were confirmed by Applied Biosystems® 7500 Real-Time PCR Systems. After the reaction, the fluorescent signal curves and the threshold line were used to interpret the mutation results.

Immunohistochemistry (IHC)
IHC was carried out using established methods. [11] In brief, sections were deparaffinized and incubated with the ALK work fluid (ALK IHC-5A4, Leica Biosystems) and ERBB2 work fluid (Her-2 IHC-UMAB36, ZSGB-BIO). A three-stage indirect immunoperoxidase technique was performed on a Benchmark Ventana staining module (Ventana, Tucson, AZ). Antigen retrieving was performed on the module using the cell conditioning buffer (CC1) pH 8.4 with Tris/Borate/EDTA (Ventana), for 1 h with the Amplification Kit (Ventana). The percentage of positive cells was evaluated, and staining scores were assessed as follows: 0, no staining; 1+, faint cytoplasmic staining; 2+, moderate cytoplasmic staining; and 3+, intense granular cytoplasmic staining.

Fluorescence in situ hybridization (FISH)
FISH was carried out using established methods. [11] Briefly, FFPE tissue sections were hybridized with probes to ERBB2 and the centromere region of chromosome 17 (CEP17) in the PathVysion ERBB2 FISH assay (Abbott-Vysis). Hybridized slides were digitally imaged and 20 no overlapping cells were evaluated for ERBB2 and CEP17 copy numbers using the Ikoniscope ERBB2 Analysis Software. ERBB2 copy number, CEP17 copy number and ERBB2:CEP17 ratio were calculated and reported according to the package insert.
ALK rearrangement status was assessed by FISH using the Vysis ALK Break Apart FISH Probe Kit, and tests were performed according to the kit instruction. In brief, slides were baked for one hour at 60°C followed by deparaffinization and rehydration. Pretreatment was performed at 80°C for 20 min, followed by protease treatment for 22 min at 37°C. The slides were dehydrated at 73°C for 3 min and incubated with probes at 37°C overnight. After washed at 75°C for 3 min, the slides were mounted with 4′,6-diamidino-2-phenylindole (DAPI) (ProLong Gold Antifade Mountant with DAPI, ThermoFisher Scientific), and analyzed under a ×60-×100 oil immersion objective using an Olympus BX-61 fluorescence microscope (Center Valley, PA). A tumor was considered ALK rearrangement positive if more than 15% of 50 (minimum) or 100 analyzed tumor cells showed split probes signals or isolated orange signals in accordance with published IASLC guidelines (IASLC Atlas of ALK Testing in Lung Cancer).

Establishment of the targeted NGS platform to detect DNA alteration using DNA samples and cancer cell lines
To establish a targeted NGS platform, we designed 16,198 DNA probes, targeting 4557 exons of 365 cancerrelated genes, 47 introns of 25 genes frequently rearranged in cancers. The capabality of the platform on DNA alteration detection was first tested in DNA samples and cancer cell lines (Fig. 1).
In DNA samples, we created DNA pools of normal cell lines from the 1000 Genomes Project, containning thousands of single nucleotide polymorphisms (SNPs) across the targeted exons that spans a broad range of MAF (5-100%) (Additional files 5 and 6: Table S5 and S6). On the other hand, 10 cancer cell lines haboring known somatic base substitutions and Indels were collected for this study. Two to ten of these 10 cell lines were randomly chosen and mixed with equal amount of DNA to form 21 pools. These 21 pools were sequenced by the targeted NGS, and the minimum sequencing coverage is 811× (Additional file 7: Table S7). 548 sites for base substitutions and 65 sites for Indels were selected for analysis across 21 pools (Additional files 8 and 9: Table S8 and S9).
In standard DNA pools, 97.5% base substitutions' MAF in cancer cell line pools are less than 30%, while 72.3% base substitutions' MAF in DNA samples are less than 30% ( Fig. 2a and b). The MAFs calculated from targeted NGS were highly consistent with the expected ones in normal DNA pools (Fig. 2c). However, the MAFs of base substitutions in cancer cell line pools determined by targeted NGS are less correlated with the expected ones (Fig. 2d), which may be due to non-diploid genomes in cancer cells. The complexity of the cancer cell lines themselves may better represent the real world clinical samples. Therefore cancer cell lines instead of the normal cell DNA could be better for the NGS platform validation.
To test the correlation between sensitivity and sequencing depth, the sequencing reads were randomly selected to in silico form a set of fastq files with sequencing depth from 0× to 800×. As expected, detection sensitivity declined with decrease of coverages, especially for those base substitutions with a MAF lower than 10% ( Fig. 2e and f ). The average detection sensitivities reach plateau at the coverage of 400× for base substitutions in normal DNA samples and at 500× in cancer cell line pools ( Fig. 2e and f ). For base substitutions with a MAF of ≤10%, 10-20% and ≥20%, the sensitivities at 500× coverage in cancer cell lines were 97.3% (1200/1233), 99.7% (1648/1653) and 100% (719/719), respectively (Fig. 2f ). These data demonstrate that the targeted NGS has high sensitivity on detection of base substitutions.
For the base substitutions with a MAF of ≤10%, 10-20% and ≥20%, the high sensitivities of detection are reached in both DNA samples and cancer cell lines at relative high coverage, while the speed of reaching the maximum in cancer cell lines is much slower than the one in DNA samples ( Fig. 2e and f ). Furthermore, the variances at different MAF groups in caner cell lines are larger than the ones in DNA samples. In other words, when the sequence depth is low, all of the real mutations can not be detected in cancer cell lines. It is suggested that the cancer cell lines can better represent the complicated features of tumor heterogeneity, and may be better for NGS platform validation.
To assess the capacity of the targeted NGS on detection of Indels, the total 1365 (MAF ≥ 1%) Indels were known in cancer cell lines and most MAF is less than 30% (Additional file 10: Fig. S1a). At 400×, the sensitivity can reach to 97.8%. Due to the complexity of the cancer cell lines, the MAFs of indels in cancer cell lines determined by targeted NGS are less correlated with the expected ones (Additional file 10: Fig. S1b). Overall, the targeted NGS has high sensitivity on detection of Indels with the relatively low sequencing depth compared with the base substitutions (Additional file 10: Fig. S1c).
To test the capacity of the targeted NGS on detection of CNV, DNA from the cell line HCC1143 with known amplifications of CCND1, FGF3, FGF4, FGF19 and AKT1, was diluted with matched normal DNA from 50% to 10%. At 50%, all known amplifications were detected, but at 20%, AKT1-amplification was undetectable (Additional file 10: Fig. S1d and S1e). It is concluded that more than 20% tumor cells in the mixed cell line pool are required to reach a high sensitivity of CNV detection. This is used as the guidline for a clinical test.

Targeted NGS to identify base substitutions and Indels from clinical specimens
To evaluate the capacity of targeted NGS to detect base substitutions and Indels from clinical specimens, we collected 34 FFPE resection specimens including 17 lung cancers, 13 colorectal cancers and 4 melanomas. Total of 12 mutation sites in four oncogenes (EGFR, KRAS, NRAS and BRAF) had been identified previously by ARMS in the hospital. Each specimen harbors at least one DNA aberration. The DNA alterations in these samples were then analyzed by targeted NGS. As was shown   Fig. 3a and Additional file 11: Table S10, all DNA aberrations were also detected by targeted NGS, indicating that the targeted NGS is as sensitive as ARMS to identify base substitutions and indels.
To further compare the detection efficiency between targeted NGS and ARMS, we gathered another batch of FFPE resection specimens, including 56 lung adenocarcinoma and 41 colorectal cancers. These specimens had been examined by targeted NGS, and 15 hotspot mutation sites in EGFR and KRAS (52 EGFR-mutated specimens,38 KRAS-mutated specimens and 7 specimens without these hot mutated sites) (Additional file 10: Fig. S2) were identified. Because of the complexity of variant in clinical tumor samples, there are a multiple of genetic muations in one tumor sample (Additional file 10: Fig. S2). Among these hotspot mutations from 97 specimens, all samples were confirmed by ARMS, and the concordance is 100% at the sample level (Fig. 3b, c and Additional file 12: Table S11). Overall, targeted NGS is as sensitive as ARMS to detect hotspot base substitutions and Indels from clinical FFPE samples.

Targeted NGS to detect DNA rearrangements from clincal specimens
We collected 18 FFPE resection specimens of lung adenocarcinoma to assess the capacity of targeted NGS to detect DNA arrangements. Among these 18 samples, 7 resection specimens had been stained by IHC and showed positive ALK immunostaining, indicating ALK fusions (Fig. 4a). These samples were re-examined by targeted NGS. The data revealed that 6 samples possessed EML4-ALK fusions, while 3D-L65 showed negative ALK fusion (Fig. 4a). Sample 3D-L65 was further investigated by IHC and FISH from the third party and confirmed to be ALK positive (Fig. 4b). The inconsistency with the targeted NGS's result could be contributed by the tumor heterogeneity or no probe coverage due to new fusion types. The remaining 11 FFPE samples were examined by targeted NGS and were positive ALK rearrangements (Fig. 4c). Among these variants of ALK fusions in this study, ZNF2-ALK had not been reported previously. All these samples were then immunostained with ALK antibody, and were identified to be ALK positive. The specificity of ALK fusion identification from targeted NGS is 100% (Fig. 4c). Therefore, targeted NGS is sensitive enough to identify ALK fusion across a large range of ALK expression level, which demonstrates high tumor heterogeneity.

Targeted NGS to identify CNVs from clinical specimens
HER2 is frequently amplified in breast cancers. [12] FISH is recognized as the "gold standard" for translocations and HER2 amplification. To investigate the detection efficiency of targeted NGS on CNVs, FISH is used to confirm HER2-amplification detected by targeted NGS. In 14 samples with the positive HER2 amplification by targeted NGS, all samples were also positively confirmed by FISH (Fig. 5a). In comparison with the golden standerd FISH, the specificity of targeted NGS was 100% (14/14  Since IHC is a widely used clinical method in China, and most breast specimens were immunostained with Her-2 antibody. The staining intensities were classified into 5 groups ranging from low to high: -,1+, 2+, 2 +~3+ and 3+, which demonstrated that clinical breast cancer samples was too complicated to be divided into positive and negtive groups. Among the 15 HER2-overexpressed samples (IHC 3+), 14 were identified to be HER2 amplification by targeted NGS (Fig. 5b). The disconcordant case, 3D-B01, was further identified to be no HER2 amplification by FISH (Fig. 5c), which suggests that overexpression of Her-2 protein in this case was not contributed by HER2 amplification. All 35 Her-2 negative (−, +) specimens were no HER2 amplification by targeted NSG (Additional file 10: Fig. S3a). In comparison with IHC 3+ results, the sensitivity and specificity of targeted NGS were 93.3% (14/15) and 100% (35/35), respectively ( Fig. 5b and Additional file 10: Fig. S3a). For the specimens with the IHC transient state (2+ and 2 +~3+), the concordance between IHC and targeted NGS is less than 50% (Additional file 10: Fig. S3a). It is suggested that high Her-2 protein expression is not only contributed by HER2 amplification.
To further elucidate the specificity of CNV identification by NGS, eight samples from multiple types of cancers have been identified to be HER2 amplification by targeted NGS. All of them have been stained by IHC and 7 out of 8 samples were scored 3+ and one scored 2+. This suggested that HER2 amplification detected by NGS leads to high Her-2 protein expression (Additional file 10: Fig. S3b).

Performance of targeted NGS on variance detection from a large non-small-cell lung cancer (NSCLC) cohort
To further prove the reliability of the targeted NGS platform on DNA alteration detection, we analyzed the spectrum of DNA alterations identified by the platform. In 215 NSCLS cases, 17 genes were identified to have the multiple types of variants (Additional file 13: Table S12). Most of them are known NSCLS driver genes, such EGFR, CDKN2A, ALK and etc. (Fig. 6a). The recurrent frequency of EGFR mutations was 56.3%, very close to the reported results in NSCLC patients in China, 46.6%~53.8%. [10] The other oncogenes also had similiar recurrent frequencies as reported in TCGA (The Cancer Genome Atlas) and other studies. [13,14] c NGS IHC(ALK) In order to investigate the detection of targeted NGS on EGFR hotspot mutations, we analyzed 121 NSCLC samples with EGFR mutations detected by targeted NGS (Additional file 14: Table S13). In these samples, the EGFR mutations associated with drug sensitivity, such as L858R and some 19 exon deletions (or insertions), were detected by targeted NGS. Moreover, the mutation rates of L858R and 19 exon deletions (or insertions) were 34% and 40% respectively (Fig. 6b), and were similar to that from other studies. [15][16][17] In addition, the EGFR mutations associated with drug resistance, such as T790 M and 20 exon mutations, [18][19][20] were also found in these NSCLC samples. In conclusion, the targeted NGS platform developed here has accurately detect EGFR hotspot mutations in NSCLC patients.

Discussion
We established a NGS platform targeting 365 cancerrelated genes to identify genomic DNA alterations including base substitutions, Indels, rearrangement and CNV. In this study, we have answered to the question -"what kind procedures need to throughly validate the new NGS platform for clinical diagonasis". This newly established NGS platform has been compared with the current clinical platforms, such as ARMS, IHC and FISH in the detection of different types of variances. Although high concordance exists between the platforms, the minor difference does demonstrate the uniqueness for each platform. For the targeted NGS, one assay can identify mutiple types of variances. The disvantage is that the turn around time (TAT) is too long for the clinical diagnosis in comparison with ARMS, IHC and FISH. If the clinical purpose is clear, the current clinical tool would be better in term of TAT and cost. This provides a clinician with the challenge to choose the right platform.
For the known hotspot variance detection like EGFR L858R and 19dels, the high concordence demonstrates that targeted NGS is as efficient as ARMS does. For those patients who do not have the hotspot mutations covered by ARMS, targeted NGS may identify the new mutations due to unbias probe design in all interesting regions, which can provide the hope for a new treatment. Specially for the relapse patient, he or she has developed resistance to tyrosine kinase inhibitors (TKIs). ALK rearrangement generates an oncogenic fusion kinase leading to ALK constitutive activation. [21,22] ALK rearrangement occurs in around 3-6% NSCLC, and is a promising therapeutic target. [23] An ALK inhibitor like crizotinib has benefit the lung adenocarcinomas patient with ALK rearrangement. [24] FISH is the only diagnostic tool approved by Food and Drug Administration (FDA) to identify ALK rearrangement. Although FISH has a high sensitivity and specificity, it can not distinguish all ALK fusion types, which are associated with the efficacy of crizotinib in patients. [25] Several studies reported that a very high concordance between IHC and FISH exists, [26,27] and IHC to determine ALK status was also approved by China Food and Drug Administration (CFDA). However, ALK antibody affinity to its fusion proteins may depend on specific variances. For example, ALK antibody CD246 from Dako only has 27% sensitivity to EML4-ALK variances 1 Fig. 6 Targeted NGS futher validated based on the spectrum of DNA alterations and EGFR hotspot mutation rates in a large non-small-cell lung cancer (NSCLC) cohort. (a) Gene alterations were found in 215 cases of NSCLC by targeted NGS. (b) Hotspot mutations in EGFR were identified by targeted NGS platform in 121 NSCLC samples and 3a/b. [28] Moreover, intracellular and extracellular mucin have effect on IHC analysis, which may cause high false-negative and false-positive detection respectively. [29] Although IHC and FISH have some disvantages, they can detect some types of ALK rearrangements that targeted NGS can not identify due to the complexity of clinical specimens. While targeted NGS is as efficient as IHC on ALK fusion detection and avoides IHC above weakness, it can also identify new ALK rearrangement.
As to HER2 amplifications, IHC is not as effective as FISH for the detection of HER2 amplification because IHC is mainly for protein expression. Protein expression level is highly correlated with amplification, but not one to one relationship exists. In this study, Her-2 IHC 3+ was highly concordant with FISH results, but IHC 3 + −2+ or 2+ showed a large discordance with the FISH results. [30] FISH has a higher predictive value than IHC for response to treatment with trastuzumab which targets Her-2. [31] Our study demonstrated that the concordance of targeted NGS in the detection of HER2 amplification was 100%, in comparison with the golden standern FISH. In light of CNV detection, targeted NGS has advantage over IHC.