Intra-arterial radioembolization with yttrium-90 microspheres ( 90Y-RE) is an increasingly used therapy for patients with unresectable liver malignancies. Over the last decade, radioactive holmium-166 poly(L-lactic acid) microspheres ( 166Ho-PLLA-MS) have been developed as a possible alternative to 90Y-RE. Next to high-energy beta-radiation, 166Ho also emits gamma-radiation, which allows for imaging by gamma scintigraphy. In addition, Ho is a highly paramagnetic element and can therefore be visualized by MRI. These imaging modalities are useful for assessment of the biodistribution, and allow dosimetry through quantitative analysis of the scintigraphic and MR images. Previous studies have demonstrated the safety of 166Ho-PLLA-MS radioembolization ( 166Ho-RE) in animals. The aim of this phase I trial is to assess the safety and toxicity profile of 166Ho-RE in patients with liver metastases.
The HEPAR study (Holmium Embolization Particles for Arterial Radiotherapy) is a non-randomized, open label, safety study. We aim to include 15 to 24 patients with liver metastases of any origin, who have chemotherapy-refractory disease and who are not amenable to surgical resection. Prior to treatment, in addition to the standard technetium-99m labelled macroaggregated albumin ( 99mTc-MAA) dose, a low radioactive safety dose of 60-mg 166Ho-PLLA-MS will be administered. Patients are treated in 4 cohorts of 3-6 patients, according to a standard dose escalation protocol (20 Gy, 40 Gy, 60 Gy, and 80 Gy, respectively). The primary objective will be to establish the maximum tolerated radiation dose of 166Ho-PLLA-MS. Secondary objectives are to assess tumour response, biodistribution, performance status, quality of life, and to compare the 166Ho-PLLA-MS safety dose and the 99mTc-MAA dose distributions with respect to the ability to accurately predict microsphere distribution.
This will be the first clinical study on 166Ho-RE. Based on preclinical studies, it is expected that 166Ho-RE has a safety and toxicity profile comparable to that of 90Y-RE. The biochemical and radionuclide characteristics of 166Ho-PLLA-MS that enable accurate dosimetry calculations and biodistribution assessment may however improve the overall safety of the procedure.
The liver is a common site of metastatic disease. Hepatic metastases can originate from a wide range of primary tumours (e.g. colorectal-, breast- and neuroendocrine tumours) . It is estimated that 50% of all patients with a primary colorectal tumour will in due course develop hepatic metastases . Once a primary malignancy has spread to the liver, the prognosis of many of these patients deteriorates significantly. Potentially curative treatment options for hepatic metastases consist of subtotal hepatectomy or, in certain cases, radiofrequency ablation. Unfortunately, only 20-30% of patients are eligible for these potentially curative treatment options, mainly because hepatic metastases are often multiple and in an advanced stage at the time of presentation . The majority of patients are therefore left with palliative treatment options.
Palliative therapy consists primarily of systemic chemotherapy. In spite of the many promising developments on cytostatic and targeted biological agents over the last ten years, there are still certain tumour types that do not respond adequately and the long-term survival rate for patients with unresectable metastatic liver disease remains low [4–8]. Moreover, systemic chemotherapy can be associated with substantial side effects that lie in the non-specific nature of this treatment. Cytostatic agents are distributed over the entire body, destroying cells that divide rapidly, both tumour cells and healthy cells. For these reasons, a significant need for new treatment options is recognized.
A relatively recently developed therapy for primary and secondary liver cancer is radioembolization with yttrium-90 microspheres ( 90Y-RE). 90Y-RE is a minimally invasive procedure during which radioactive microspheres are instilled selectively into the hepatic artery using a catheter. The high-energy beta-radiation emitting microspheres subsequently strand in the arterioles (mainly) of the tumour, and a tumoricidal radiation absorbed dose is delivered. The clinical results of this form of internal radiation therapy are promising [9, 10]. The only currently clinically available microspheres for radioembolization loaded with 90Y are made of either glass (TheraSphere ®, MDS Nordion Inc., Kanata, Ontario Canada) or resin (SIR-Spheres ®, SIRTeX Medical Ltd., Sydney, New South Wales, Australia).
Although 90Y-RE is evermore used and considered a safe and effective treatment, 90Y-MS have a drawback: following administration the actual biodistribution cannot be accurately visualized. For this reason, holmium-166 loaded poly(L-lactic acid) microspheres ( 166Ho-PLLA-MS) have been developed at our centre [11, 12]. Like 90Y, 166Ho emits high-energy beta particles to eradicate tumour cells but 166Ho also emits low-energy (81 keV) gamma photons which allows for nuclear imaging. As a consequence, visualization of the microspheres is feasible. This is very useful for three main reasons. Firstly, prior to administration of the treatment dose, a small scout dose of 166Ho-PLLA-MS can be administered for prediction of the distribution of the treatment dose. This provides a theoretical advantage over 90Y-RE, for which the distribution assessment depends on a scout dose of 99mTc-MAA, with a disputable distribution correlation with the actual microspheres . Secondly, quantitative analysis of the nuclear images would allow assessment of the radiation dose delivered on both the tumour and the normal liver (i.e. dosimetry) . Thirdly, since holmium is highly paramagnetic, it can be visualized using magnetic resonance imaging (MRI). Quantitative analysis of these MRI images is also possible, which is especially useful for medium- and long-term monitoring of the intrahepatic behaviour of the microspheres [15, 16].
The pharmaceutical quality of 166Ho-PLLA-MS has been thoroughly investigated and proven to be satisfactory [17–19]. Multiple animal studies have been conducted in order to investigate the intrahepatic distribution (ratio tumour to normal liver), the toxicity profile/biocompatibility of the 166Ho-PLLA-MS, safety of the administration procedure, and efficacy of these particles [20–23].
Now that the preclinical phase of 166Ho-RE has been successfully completed, we will start a clinical trial (the HEPAR study: H olmium E mbolization P articles for A rterial R adiotherapy) in order to evaluate 166Ho-RE in patients with liver metastases. The main purpose of this trial is to assess the safety and toxicity profile of 166Ho-RE. Secondary endpoints are tumour response, biodistribution prediction with 99mTc-MAA versus a safety dose of 166Ho-PLLA-MS, performance status, and quality of life.
The HEPAR study is a single centre, non-randomized, open label safety study. In this phase I study, a new device will be investigated, namely 166Ho-PLLA-MS for intra-arterial radioembolisation for the treatment of liver malignancies. In a group of 15 to 24 patients with liver metastases, treated with increasing amounts of 166Ho, the device will be investigated for safety and toxicity.
The study will include patients with liver-dominant metastases, of any histology, who cannot be treated by standard treatment options such as surgery and systemic chemotherapy, due to advanced stage of disease, significant side effects or unsatisfactory tumour response. The detailed inclusion and exclusion criteria are listed in Appendix 1.
Patient recruitment will take place between October 2009 and January 2011.
Using the solvent evaporation technique, non-radioactive holmium-165 ( 165Ho) and its acetylacetonate complex (HoAcAc) can be incorporated into the poly(L-lactic acid) matrix to form microspheres (Figure 1). Subsequently, the non-radioactive 165Ho-PLLA-MS can be made radioactive by neutron activation in a nuclear facility and form 166Ho-PLLA-MS. Neutron-activated 166Ho has a half-life of 26.8 hours and is a beta emitter (Eβmax = 1.85 MeV) that also emits gamma photons (Eγ = 81 keV) suitable for single photon emission computed tomography (SPECT) (Table 1).
Patients with liver metastases who agree to participate in the study must be referred to the principle investigator by the department of Surgery. The principle investigator will inform every patient and obtain their informed consent.
A screening visit will take place at the outpatient clinic within 14 days prior to the fist angiography. During this visit, the principle investigator will run through the inclusion and exclusion criteria, conduct a physical examination, and assess the WHO performance status of the patient. Subsequently, CT, MRI, and positron emission tomography (PET) will be performed, as well as electrocardiography (ECG). PET will only be performed in FDG-avid tumours. Liver weight will be calculated, based on the liver volume measured on CT data with a density conversion factor of 1.0 g/cm 3. Relevant laboratory tests (haematology, coagulation profile, serum chemistry, tumour marker) must be documented and reviewed. All patients are asked to fill out the European Organisation for Research and Treatment of Cancer (EORTC) QLQ-C30 questionnaire .
Patients will be hospitalized on the evening prior to angiography. On day 0 the patient is subjected to angiography of the upper abdominal vessels. The celiac axis and superior mesenteric artery are visualised, followed by coiling of relevant vessels, in particular branches of the hepatic artery supplying organs other than the liver, e.g. gastroduodenal artery (GDA), right gastric artery (RGA). If major arteries like the GDA or RGA cannot be successfully occluded, the patient will be withheld 166Ho-RE. This procedure will be performed by a skilled and trained interventional radiologist. The catheter is introduced using the Seldinger technique. Prior to the procedure, the patient is offered a tranquilizer (oxazepam 1 dd 10 mg). Premedication consists of a single administration of corticosteroids (dexamethason 10 mg i.v.) and antiemetics (ondansetron 8 mg i.v.). Proton pump inhibitors (pantoprazol 1 dd 40 mg) are started on the day of the intervention and prescribed for use until the end of the follow-up.
Macroaggregated albumin injection
After successful angiography and coiling of relevant vasculature is performed, a dose of 99mTc-Macroaggregated Albumin ( 99mTc-MAA) will be administered in the hepatic artery on the same day. The 99mTc-MAA are used to assess whether a favourable distribution of the 166Ho-PLLA-MS can be expected. The patient is subjected to planar imaging of the thorax and abdomen and SPECT of the abdomen, in order to determine the 99mTc-MAA distribution. Images will be evaluated qualitatively and quantitatively. Extrahepatic deposition of activity is a contra-indication for administration of the treatment dose. Region of interest analysis will be used to calculate lung shunting. Lung shunting should not exceed 20% of the dose 99mTc-MAA. If the amount of lung shunting cannot be reduced to <20% using standard radiological interventional techniques to decrease the shunting, the patient will not be eligible to receive a safety nor a treatment dose of 166Ho-PLLA-MS. The dose point-kernel method will be applied to the (non-homogeneous) activity distribution to calculate the absorbed dose distribution . Dose-volume histograms will be generated in order to quantify the dose distribution, and the tumour to healthy tissue absorbed dose ratio will be calculated.
166Ho-PLLA-MS safety dose
The second angiography takes place around 1 week after the first angiography but no longer than 2 weeks later. Patients will be hospitalized on the evening before the day of treatment. They will be discharged approximately 48 hours after the intervention unless complications have occurred. Prior to the procedure, the patient is offered a tranquilizer (oxazepam 10 mg). A safety dose of 166Ho-PLLA-MS will be administered through a catheter inside the hepatic artery, at the position planned during the first intervention. The safety dose will consist of 60 mg (10% of the total amount of microspheres) 166HoPLLA-MS with a lower specific activity (90 Bq/microsphere) than for the treatment dose. After the safety dose, planar imaging of both the thorax and abdomen will be performed, as well as SPECT and MRI of the abdomen. Presence of inadvertent administration to the lungs or other upper abdominal organs will once more be checked for. These SPECT and MRI images will be compared with the images post 99mTc-MAA and post-treatment, regarding extrahepatic deposition of activity, percentage lung shunting, homogeneity of the dose distribution and tumour to healthy tissue absorbed dose ratio.
166Ho-PLLA-MS treatment dose
When the amount of lung shunting does not exceed 20% of the safety dose of 166HoPLLA-MS, the (complete) treatment dose of 166HoPLLA-MS will be administered (Figure 2). Consecutive cohorts of 3 patients will be treated with identical amounts of microspheres (600 mg), and the last cohort will consist of at least 6 patients. If no toxicity ≥ grade 3 according to the Common Terminology Criteria for Adverse Events (CTCAE) is observed, the next cohort of three patients will be treated at the next radiation dose level. If in one patient CTCAE ≥ grade 3 is observed in a particular cohort, the cohort will be extended to six patients. If toxicity ≥ grade 3 is observed in two or more patients in a particular cohort, the study will be terminated because the endpoint, e.g. the maximum tolerated radiation dose, is reached. This will be reported to the Independent Ethics Committee (IEC). The dose level prior to the toxic radiation dose will become the recommended dose for efficacy studies. If an event is classified as grade 3 or 4 administration technique related, the patient will be replaced. The specific activity of the 166Ho-PLLA-MS will be increased by adapting the activation time in the nuclear reactor. The first, second, third and fourth cohort will be treated with a dose of 1.3, 2.5, 3.8 and 5.0 GBq/kg (liver weight), respectively. Assuming a homogenous uptake throughout the liver, this equals escalating radiation doses of 20 Gy, 40 Gy and 60 Gy, to a maximum dose of 80 Gy in the last cohort. A maximum of 15.1 GBq will be given to the maximum treated liver weight (inclusive the tumour tissue) of 3 kg (Table 2). The amount of radioactivity administered to the patient is calculated according to the following formula:
where LW is the liver weight of the patient which may be determined using CT, MRI or ultrasound, and where 15.87 × 10 -3 (J/MBq) is the activity-to-dose conversion factor for 166Ho .
Radiation exposure rate
During the hospitalization in week 1 the radiation exposure rate will be measured from 1 m distance at t = 0, 3, 6, 24, and 48 hours following 166Ho-PLLA-MS administration. Patients will not be discharged from the hospital until the dose equivalent is less than 90 μSv/h measured from 1 m distance.
All patients are followed over a period of 12 weeks after treatment with weekly visits at the outpatient clinic. During each visit, data is collected by physical examination, WHO performance status assessment and laboratory examination (haematology, coagulation profile, serum chemistry and (if applicable) tumour marker). Adverse events are monitored. In addition, patients are asked to fill out the EORTC questionnaires in the 6 th and 12 th week post-treatment. CT and (in case of 18F-FDG-avid tumours) PET are performed in the 6 th and 12 th week post-treatment and MRI is performed in the 1 st and the 12 th week post-treatment (Figure 3).
Pooled urine samples will be collected from 0-3 hours, 3-6 hours, 6-24 hours and 24-48 hours post- 166Ho-PLLA-MS administration. In the 6 th and 12 th week post treatment, pooled 24-hours urine will be collected for measurement of holmium content. The date and time of the start and the end of the collection period, the volume and whether the collection was complete or not, will be noted in the case record form. During the hospitalization in week 1, blood will be drawn for measuring the holmium content in the blood at t = 0, 3, 6, 24, and 48 hours following 166Ho-PLLA-MS administration. Measurements will be done according to activity measurement of holmium-166 metastable ( 166mHo, T 1/2 ≈ 1200 year) with a low-background gamma-counter (Tobor, Nuclear Chicago, Chicago, IL, USA) as previously described in one of the preclinical studies by Zielhuis et al. .
The primary objective of this study is to establish the safety and toxicity profile of treatment with 166Ho-PLLA-MS. This profile will be established using the CTCAE v3.0 methodology and will be used to determine the maximum tolerated radiation dose. Any of the following events which are considered possibly or probably related to the administration of 166Ho-PLLA-MS will be considered a serious adverse event during the 12 weeks follow-up period:
Any other grade 3 or 4 toxicity (excluding expected AST/SGOT, ALT/SGPT elevation, elevated bilirubin and lymphopenia) possibly related to study device, using CTCAE v3.0.
Any life threatening event possibly related to the study device: events as a consequence of inadvertent delivery of 166Ho-PLLA-MS into non-target organs like the lung (radiation pneumonitis), the stomach and duodenum (gastric/duodenal ulcer or perforation), the pancreas (radiation pancreatitis), and liver toxicity due to an excessive radiation dose ("radiation induced liver disease" (RILD) ).
The haematological and biochemical adverse events as well as RILD will be considered dose limiting toxicity.
Secondary objectives are to evaluate tumour response, performance status, biodistribution, quality of life and to compare the accuracy of the 99mTc-MAA scout dose with a safety dose of 166Ho-PLLA-MS, in predicting microsphere distribution of the treatment dose. Tumour response will be quantified using CT of the liver scored according to Response Evaluation Criteria in Solid Tumours guidelines (RECIST 1.1) . Tumour viability will be assessed by PET, depending on tumour type. In addition, the antitumoral effect will be assessed by relevant tumour markers responses if applicable (i.e. carcinoembryonic antigen (CEA) in colorectal carcinoma and chromogranin A (CgA) for neuroendocrine tumours). Biodistribution is assessed using quantitative SPECT and MRI. Urine and blood samples will be screened for presence of 166Ho-PLLA-MS or fragments of 166Ho-PLLA-MS. Performance status is assessed using WHO performance status criteria. Quality of life (QoL) is evaluated using the EORTC questionnaire QLQ-C30 with colorectal liver metastases module QLQ-LMC21. Finally, the accuracy of the 166Ho-PLLA-MS safety dose in predicting the distribution of the treatment dose is compared with the accuracy of the 99mTc-MAA. Quantitative SPECT analysis will be performed using the scatter correction method described by De Wit et al. .
From the literature on 90Y-RE, it is known that several treatment related effects can occur in radioembolization. As long as the patient is treated with the correct technique, which includes that no excessive radiation dose be delivered to any organ, the common adverse events after receiving radioactive microspheres are fever, abdominal pain, nausea, vomiting, diarrhoea and fatigue (i.e. postembolization syndrome) [10, 28–30]. These effects are in general self-limiting within 1 to 2 weeks, and may be up to grade 3 or 4 (CTCAE v3.0) without direct clinical relevance. Based on the preclinical studies, a similar safety profile is expected for 166Ho-RE [22, 23].
Patients will receive oral analgesics (paracetamol up to 4000 mg/24 h) for relief of fever and pain after the administration of microspheres. To reduce nausea and vomiting, patients will receive anti-emetics (ondansetron up to 3 dd 8 mg) during the first 24 hours after administration of the treatment dose. In the case of persisting nausea, metoclopramid (up to 300 mg/24 h) will be used. Patients suffering from diarrhoea will receive loperamide (up to 16 mg/24 h). The vascular contrast agent jodixanol (Visipaque ®) may cause renal insufficiency in poorly hydrated patients. All patients will therefore be hydrated. This consists of 1.5 l NaCl 0.9% both prior to and post angiography. Inadvertent delivery of microspheres into organs such as the lungs, stomach, duodenum, pancreas, and gallbladder is associated with serious side effects. To reduce toxicity of the radioactive microspheres in patients with excessive extrahepatic deposition of 166Ho-PLLA-MS, the cytoprotective agent amifostine (Ethyol ®, up to 200 mg/m 2 for 7 days) may be administered intravenously.
Descriptive statistics (n, mean, standard deviation, median, minimum and maximum) will be calculated for each quantitative variable; frequency counts by category will be made for each qualitative variable. Interim analysis will be performed after every 3 patients. Inclusion of patients in the next cohort will be performed if the Independent Data Monitoring Committee (IDMC) has scrutinized the toxicity data and given permission to proceed.
Two sets of study data will be evaluated: the primary objective will be evaluated in the full analysis set (FAS). The FAS is defined as the set of data generated from the included patients who received at least the safety dose. The secondary objectives will be evaluated in both FAS and per-protocol set (PPS). The PPS is defined as the set of data generated from the included patients who complied with the protocol.
The IDMC will perform a safety review after each series of treatments of three consecutive patients. The IDMC members have no conflict of interest with the sponsor because they are not involved in the study, nor are they receiving funds. The IDMC will work according to standard operating procedures and will receive reports on a regular basis on all toxicity CTCAE ≥ grade 3 reported for this trial. Recruitment will not be interrupted unless otherwise requested by the chairman of the IDMC. The responsibilities of the IDMC include:
minimize the exposure of patients to an unsafe therapy or dose
make recommendations for changes in study processes where appropriate
endorse continuation of the study
inform the institutional IEC in the case of toxicity CTCAE ≥ grade 3 and/or when the well-being of the subjects is jeopardized
The study will be conducted according to the principles of the Declaration of Helsinki (version 9.10.2004) and in accordance with the Medical Research Involving Human Patients Act (WMO), the requirements of International Conference on Harmonization - Good Clinical Practice. The study protocol has been approved by the IEC and by the institutional Radiation Protection Committee.
The HEPAR trial is a phase I study to evaluate the safety and toxicity profile of 166Ho radioembolization. Secondary endpoints are tumour response, biodistribution assessment, performance status, quality of life and comparison of the biodistributions of the 99mTc-MAA scout dose and the 166Ho-PLLA-MS safety dose.
With regard to the method of administration, viz. through a catheter placed in the hepatic artery, the in-vivo characteristics (no significant release of radionuclide), and the mechanism of action (local irradiation of the tumour), 166Ho-PLLA-MS constitute a device analogous to the 90Y microspheres, which are currently applied clinically. 166Ho-PLLA-MS only differ in the radioisotope and the device matrix that are used. In a toxicity study in pigs on 166Ho-RE, it has been demonstrated that (healthy) pigs can withstand extremely high liver absorbed doses, at least up to 160 Gy . During these animal experiments, only very mild side effects were seen: slight and transitory inappetence and somnolence, which may well have been associated with the anaesthetic and analgesic agents that had been given and not necessarily with the microsphere administration. It is plausible that this low toxicity profile is caused by the inhomogeneous distribution of 166Ho within the liver after intra-arterial injection, as was observed on MRI and SPECT images. The current study will investigate whether a similar distribution pattern can also be observed in human subjects and whether this inhomogeneous distribution is concentrated around the tumour sites.
Hepatic arterial injection with 99mTc-MAA and subsequent scintigraphic imaging is widely used to predict the biodistribution of 90Y microspheres, prior to the actual radioembolization procedure. Its accuracy can however be disputed. In our centre, we have observed that patients with a borderline lung shunt fraction of 10% to 19%, as calculated using the 99mTc-MAA images (approximately 24% of all patients, all of whom were instilled a by 50% reduced amount of radioactivity), had no signs of lung shunting on post- 90Y-RE Bremsstrahlung images. In these cases, it seems that the 99mTc-MAA-scan had false-positively predicted extrahepatic spread. This may be explained by the fact that 99mTc-MAA differs in many aspects from the microspheres that are used. Shape, size, density, in-vivo half-life, and number of 99mTc-MAA particles do not resemble the microspheres in any way [13, 31]. In addition, free technetium that is released from the MAA particles can disturb the (correct) assessment of extrahepatic spread. We hypothesize that a small safety dose with low-activity 166Ho-PLLA-MS will be a more accurate predictor of distribution than 99mTc-MAA. The unique characteristics of 166Ho-microspheres, in theory, allow a more accurate prediction of the distribution with the use of scintigraphy and MRI. In this study, we chose to perform both an injection with 99mTc-MAA and administration of a safety dose of 166Ho-PLLA-MS. The respective distributions of the 99mTc-MAA and the 166Ho-PLLA-MS safety dose will be compared with the distribution of the treatment dose of 166Ho-PLLA-MS by quantitative analysis of the scintigraphic images.
Both commercially available 90Y-MS products are approved by the Food and Drug Administration (FDA) and European Medicines Agency as a medical device and not as a drug. Radioactive microspheres are a medical device since these implants do not achieve any of their primary intended purposes through chemical action within or on the body and are not dependent upon being metabolized for the achievement of their primary intended purpose. In accordance with the definition of a medical device by the FDA and in analogy with the 90Y-MS, we consider the 166Ho-PLLA-MS to be a medical device . The Dutch medicine evaluation board has discussed this issue (13 July 2007) and has concluded that the microspheres are indeed to be considered as a medical device.
One important issue concerning the resin-based SIR-Spheres ® is the relatively high number of particles instilled (>1,000 mg), since this may sometimes be associated with macroscopic embolization as observed during the fluoroscopic guidance [28, 33]. Several authors have reported stasis of flow during administration of resin microspheres and were forced to end the procedure prematurely because of the risk of backflow, hence extrahepatic deposition of a part of the dosage [28, 34, 35]. The specific activity of the 166Ho-PLLA-MS is considerably higher than that of the resin microspheres (≤450 and 50 Bq/microspheres, respectively). However, in order to obtain an equivalent absorbed dose, the total amount of radioactivity of the administered microspheres in 166Ho radioembolization needs to be 3 times higher than in 90Y radioembolization, due to the shorter physical half-life of 166Ho. Even so, compared with the resin 90Y microspheres, in 166Ho radioembolization considerably less microspheres (≤600 mg) are used to obtain an equivalent radiation dose, resulting in a lower risk of stasis or backflow during administration [9, 29]. A further issue is that 90Y microspheres can not be visualized under fluoroscopy during injection. Manufacturers of resin 90Y microspheres state that their microspheres are to be administered with water for injection alternated with non-ionogenic contrast . As a result, the operating physician cannot detect stasis or backflow of microspheres until he has switched from injecting microspheres to injecting the contrast agent. Holmium microspheres, on the contrary, are administered in a mixture of 50% saline and 50% non-ionogenic contrast under constant fluoroscopic imaging, which ensures constant control over the microspheres during injection . However, continuous fluoroscopic imaging during microsphere administration may comprise an increased radiation dose delivered to the patient, specifically the abdominal skin, during the procedure.
If this phase I trial provides sufficient data to prove that 166Ho-PLLA-RE has an acceptable safety and toxicity profile, further studies will be needed. The next step will be an efficacy study in a larger number of patients. The primary endpoints of that study will be tumour response and survival.
Appendix 1 - Eligibility criteria for 166Ho-RE
Signed informed consent letter
Age >18 years
Liver-dominant metastases without standard treatment options. Liver-dominant disease is defined as the diameter of all metastases in the liver to be more than 200% of the sum of the diameters of all soft tissue lesions outside the liver.
Life expectancy of ≥12 weeks
World Health Organisation (WHO) Performance status 0-2
≥1 measurable lesions of ≥10 mm in the longest diameter by spiral computed tomography (CT) (5 mm slice thickness)
Negative pregnancy test for women
Brain metastases or spinal cord compression, unless irradiated at least 4 weeks prior to the date of the experimental treatment, and stable without steroid treatment for at least 1 week
Radiation therapy within the last 4 weeks before study enrolment
Patient has received chemotherapy within 4 weeks prior to enrolment
Major surgery within 4 weeks, or incompletely healed surgical incision before enrolment
Any unresolved toxicity greater than National Cancer Institute (NCI), Common Terminology Criteria for Adverse Events (CTCAE version 3.0) grade 2 from previous anti-cancer therapy
Alanine aminotransferase (ALT), aspartate aminotransferase (AST), or alkaline phosphatase (ALP) >5× Upper Limit of Normal (ULN), serum bilirubin >1.5× ULN or serum creatinine >185 µmol/L
Significant cardiac event (e.g. myocardial infarction, superior vena cava (SVC) syndrome, New York Heart Association (NYHA) classification of heart disease ≥2 within 3 months before entry, or presence of cardiac disease that, in the opinion of the investigator, increases the risk of ventricular arrhythmia
Pregnancy or breast feeding
Comorbidity with a grave prognosis (estimated survival <3 months) and/or worse than the basic disease for which the patients will be included in the study
Abnormalities of the bile ducts (such as stents) with an increased chance of infection
Diseases with an increased chance of liver toxicity, such as primary biliary cirrhosis or xeroderma pigmentosum
Patients who are declared incompetent or have a psychiatric disorder that makes a comprehensive judgement impossible, such as psychosis, hallucinations and/or depression
Previous enrolment in the present study or previous treatment with radioembolization
Treatment with an investigational agent within 42 days prior to enrolment
Female patients who are not using an acceptable method of contraception or are less than 1 year postmenopausal or surgically sterile during their participation in this study (from the time the consent form is signed) to prevent pregnancy
Male patients who are not surgically sterile or do not use an acceptable method of contraception during their participation in this study to prevent pregnancy in a partner
Evidence of portal hypertension, splenomegaly or ascites
Body weight >150 kg
Active hepatitis (B and/or C)
Liver weight >3 kg (determined by software using CT data)
Allergy for intravenous contrast agent used (Visipaque ®)
General MRI contra-indications (severe claustrophobia, metal implants, implanted pacemaker and/or neurostimulators)
Patients who have arterial variations that will not allow whole liver treatment by a single administration via the hepatic artery
Russell AH, Tong D, Dawson LE, Wisbeck W: Adenocarcinoma of the proximal colon. Sites of initial dissemination and patterns of recurrence following surgery alone. Cancer. 1984, 53: 360-367. 10.1002/1097-0142(19840115)53:2<360::AID-CNCR2820530232>3.0.CO;2-U.
Vente MAD, Hobbelink MGG, Van het Schip AD, Zonnenberg BA, Nijsen JFW: Radionuclide liver cancer therapies: from concept to current clinical status. Anticancer Agents Med Chem. 2007, 7: 441-459. 10.2174/187152007781058569.
Salem R, Thurston KG: Radioembolization with yttrium-90 microspheres: a state-of-the-art brachytherapy treatment for primary and secondary liver malignancies: part 3: comprehensive literature review and future direction. J Vasc Interv Radiol. 2006, 17: 1571-1593. 10.1097/01.RVI.0000236744.34720.73.
Nijsen JFW, Van Steenbergen MJ, Kooijman H, Talsma H, Kroon-Batenburg LM, Van de Weert M, Van Rijk PP, De Witte A, Van het Schip AD, Hennink WE: Characterization of poly(L-lactic acid) microspheres loaded with holmium acetylacetonate. Biomaterials. 2001, 22: 3073-3081. 10.1016/S0142-9612(01)00055-2.
Seppenwoolde JH, Nijsen JFW, Bartels LW, Zielhuis SW, Van het Schip AD, Bakker CJ: Internal radiation therapy of liver tumors: Qualitative and quantitative magnetic resonance imaging of the biodistribution of holmium-loaded microspheres in animal models. Magn Reson Med. 2004, 53: 76-84. 10.1002/mrm.20320.
Zielhuis SW, Nijsen JFW, De Roos R, Krijger GC, Van Rijk PP, Hennink WE, Van het Schip AD: Production of GMP-grade radioactive holmium loaded poly(l-lactic acid) microspheres for clinical application. Int J Pharm. 2006, 311: 69-74. 10.1016/j.ijpharm.2005.12.034.
Zielhuis SW, Nijsen JFW, Dorland L, Krijger GC, Van het Schip AD, Hennink WE: Removal of chloroform from biodegradable therapeutic microspheres by radiolysis. Int J Pharm. 2006, 315: 67-74. 10.1016/j.ijpharm.2006.02.010.
Aaronson NK, Ahmedzai S, Berman B, Bullinger M, Cull A, Duez NJ, Filiberti A, Flechtner H, Fleishman SB, De Haes JC, et al: The European Organization for Research and Treatment of Cancer QLQ-C30: a quality-of-life instrument for use in international clinical trials in oncology. J Natl Cancer Inst. 1993, 85: 365-376. 10.1093/jnci/85.5.365.
Trotti A, Colevas AD, Setser A, Rusch V, Jaques D, Budach V, Langer C, Murphy B, Cumberlin R, Coleman CN: CTCAE v3.0: development of a comprehensive grading system for the adverse effects of cancer treatment. Semin Radiat Oncol. 2003, 13: 176-181. 10.1016/S1053-4296(03)00031-6.
Goin J, Dancey JE, Roberts C: Comparison of post-embolization syndrome in the treatment of patients with unresectable hepatocellular carcinoma: Trans-catheter arterial chemo-embolization versus yttrium glass microspheres. World J Nucl Med. 2004, 3: 49-56.
Vente MA, Wondergem M, Van der Tweel I, Van den Bosch MA, Zonnenberg BA, Lam MG, Van het Schip AD, Nijsen JF: Yttrium-90 microsphere radioembolization for the treatment of liver malignancies: a structured meta-analysis. Eur Radiol. 2009, 19: 951-959. 10.1007/s00330-008-1211-7.
SIRTeX medical training manual: SIRTeX medical training manual. TRN-US-03, 40
Vente MA, De Wit TC, Van den Bosch MA, Bult W, Seevinck PR, Zonnenberg BA, De Jong HW, Krijger GC, Bakker CJ, Van het Schip AD: Holmium-166 poly(L -lactic acid) microsphere radioembolisation of the liver: technical aspects studied in a large animal model. Eur Radiol. 2010, 20: 862-869. 10.1007/s00330-009-1613-1.
The authors thank Ms. Tjitske Bosma (clinical research coordinator, University Medical Center Utrecht) for her contribution to the study design and coordination, and Mr. Remmert de Roos for his assistance in the preparation of the microspheres. This study was financially supported by the Dutch Cancer Society (KWF Kankerbestrijding), under grant UU2009-4346.
Authors and Affiliations
Department of Radiology and Nuclear Medicine, University Medical Center Utrecht, Heidelberglaan 100, E01.132, 3584, CX, Utrecht, The Netherlands
Maarten LJ Smits, Johannes FW Nijsen, Maurice AAJ van den Bosch, Marnix GEH Lam, Maarten AD Vente, Julia E Huijbregts, Alfred D van het Schip, Mattijs Elschot, Wouter Bult, Hugo WAM de Jong & Bernard A Zonnenberg
Department of Clinical Pharmacy, University Medical Center Utrecht, Heidelberglaan 100, 3584, CX, Utrecht, The Netherlands
The authors declare that they have no competing interests.
All authors contributed to the study design. BZ is the study's principal investigator. The manuscript was written by MS, JN, MvdB, ML, MV, and AvhS. All authors revised the manuscript and approved the final version of the manuscript.
Authors’ original submitted files for images
Below are the links to the authors’ original submitted files for images.
This article is published under license to BioMed Central Ltd. This is an Open Access article is distributed under the terms of the Creative Commons Attribution License (
), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Smits, M.L., Nijsen, J.F., van den Bosch, M.A. et al. Holmium-166 radioembolization for the treatment of patients with liver metastases: design of the phase I HEPAR trial.
J Exp Clin Cancer Res29, 70 (2010). https://doi.org/10.1186/1756-9966-29-70