Innovation in catheter design for intra-arterial liver cancer treatments results in favorable particle-fluid dynamics
© van den Hoven et al. 2015
Received: 1 June 2015
Accepted: 6 July 2015
Published: 1 August 2015
Liver tumors are increasingly treated with radioembolization. Here, we present first evidence of catheter design effect on particle-fluid dynamics and downstream branch targeting during microsphere administrations.
Materials and methods
A total of 7 experiments were performed in a bench-top model of the hepatic arterial vasculature with recreated hemodynamics. Fluorescent microspheres and clinically used holmium microspheres were administered with a standard microcatheter (SMC) and an anti-reflux catheter (ARC) positioned at the same level along the longitudinal vessel axis. Catheter-related particle flow dynamics were analyzed by reviewing video recordings of UV-light illuminated fluorescent microsphere administrations. Downstream branch distribution was analyzed by quantification of collected microspheres in separate filters for two first-order branches. Mean deviation from a perfectly homogenous distribution (DHD) was used to compare the distribution homogeneity between catheter types.
The SMC administrations demonstrated a random off-centered catheter position (in 71 % of experiments), and a laminar particle flow pattern with an inhomogeneous downstream branch distribution, dependent on catheter position and injection force. The ARC administrations demonstrated a fixed centro-luminal catheter position, and a turbulent particle flow pattern with a more consistent and homogenous downstream branch distribution. Quantitative analyses confirmed a significantly more homogeneous distribution with the ARC; the mean DHD was 40.85 % (IQR 22.76 %) for the SMC and 15.54 % (IQR 6.46 %) for the ARC (p = 0.047).
Catheter type has a significant impact on microsphere administrations in an in-vitro hepatic arterial model. A within-patient randomized controlled trial has been initiated to investigate clinical catheter-related effects during radioembolization treatment.
KeywordsRadioembolization Particle-fluid dynamics Catheter design Vascular model Liver tumors
Primary liver tumors (hepatocellular carcinoma, intrahepatic cholangiocarcinoma) and liver metastases affect a great number of cancer patients worldwide, and advanced disease stages are generally associated with poor prognosis [1–4]. Only a minority of patients are eligible for potentially curative surgery, and the efficacy of palliative systemic therapy is dependent on tumor type [5–8]. In the last decade, image-guided treatment techniques have evolved as another therapeutic option, and minimally invasive, trans-catheter treatments such as transarterial chemoembolization and intra-arterial radioembolization have found their way to patients with irresectable and chemorefractory disease [9, 10].
During radioembolization treatment, radioactive microspheres are administered through a catheter placed in the hepatic arterial vasculature. Since liver tumors are almost exclusively vascularized by the hepatic artery, and healthy liver tissue receives the majority of its blood supply from the portal vein, arterial blood flow should preferentially transport the microspheres towards tumorous tissue, where they lodge in the distal vessels surrounding tumors and emit tumoricidal high-energy β-radiation, while relatively sparing healthy liver tissue . Yet, individual tumors may receive sub-therapeutic doses of radioactivity as a result of the complex interplay between tumor vascularization, particle-fluid dynamics and catheter placement [12–14].
So far, the standard end-hole microcatheter (SMC) has been the undisputed device of choice for microsphere administrations. However, a novel catheter type has recently been developed specifically for trans-catheter liver cancer treatments. This anti-reflux catheter (Surefire Infusion System, Surefire Medical Inc., Westminster, Co, USA) features a dynamically expandable tip that prevents reflux of particles in reverse flow conditions, while preserving normal antegrade blood flow . In addition, the catheter orifice is fixed in the center of the vessel lumen (centro-luminal position). These marked differences may affect fluid-particle dynamics during microsphere administrations, and have a significant impact on tumor targeting during radioembolization.
Remarkably, fluid-particle dynamics are still inadequately understood, and the potential effects of catheter design and position remain unknown. The aim of this study is to enhance our understanding of this complex interplay by comparison of microsphere administrations with an ARC and SMC in the controlled environment of an experimental vascular model. In addition, we elaborate on potential clinical benefits that may be associated with favorable catheter characteristics.
Description of experiments
RHA, proximal to S4A
RHA, proximal to S4A
RHA, proximal to S4A
RHA, proximal to S4A
In vitro hepatic arterial model
A 200 mg dose of fluorescent red polyethylene microspheres (Cospheric LLC, Santa Barbara, Ca, USA) with a density of 1.20 g/cc and size of 27–32 μm was used for experiments 1–3. Fluorescent illumination of these microspheres is induced at a wavelength of 300–550 nm. 600 mg non-irradiated, 165Ho loaded poly (L-lactic acid) microspheres (QuiremSpheres, Quirem Medical Inc., Utrecht, The Netherlands) with a density of 1.40 g/cc and size of 30 μm (range 20–50 μm) were used for experiments 4–7. Both microsphere types were prepared for administration according to manufacturers instructions. The fluorescent microspheres were suspended in a tween surfactant solution (Tween 20) to prevent aggregation, and administered with 30 ml of water, whereas the holmium microspheres were suspended in an isotonic phosphate buffer, and administered with 30 ml of 0.9 % NaCl solution.
For the first experiment only, microspheres were injected by manual operation of a 30 cc syringe, using an intermittent short pulse pattern, similar to clinical practice. For a brief period of time, injection force was varied from minimal (pulse injection rate 0.3 ml/sec), to nominal (0.6 ml/sec), and ultimately excessive (1.2 ml/sec), in order to evaluate the impact of injection force on downstream branch distribution. The time points of different injection force were recorded. During the other experiments (experiments 2–7), an automatic syringe pump was used with standardized settings for injection (pulses of 0.5 ml, 1 sec interval between pulses, 5 ml/min) to eliminate all variability for an unbiased comparison between SMC and ARC administrations. Intra-vascular flow continued for 15 min after the beginning of administration.
Qualitative analysis of catheter-related flow patterns
All fluorescent microsphere administrations were filmed using a mounted high definition camera, under UV-light illumination to assess catheter-related flow patterns qualitatively. The video recordings of all fluorescent microsphere administrations with the SMC and ARC were reviewed side-by-side to facilitate the comparison of particle flow patterns for each catheter type.
Quantitative analysis of microsphere distribution
The distal vessels of the first-order branches (LHA/RHA for experiments 1–3, and S4A/RHA for experiments 4–7) drained into separate filters of Dutch-weave stainless steel with a rating of 10 μm (validated in-house for successful filtering of microspheres). These filters were weighed on a precision scale (AB135-S/FACT laboratory scale, Mettler-Toledo Inc., Columbus, Ohio, USA) prior to administration, and again post-administration after 12 h of heating at 190-250 °F in a digitally controlled Thermolyne 1500 furnace (ThermoFisher Scientific, Waltham, Mass., USA) to evaporate the residual water and glycerin. The measured change in mass was used to quantify the microspheres collected. The distribution over the main branches was subsequently determined by calculating the proportional weight change per filter as a function of the total weight change for both filters. In order to compare the distribution homogeneity across catheter types, the deviation from a perfect homogenous distribution (50:50 %), abbreviated as DHD, was calculated in percentage points (for example, a 10:90 % distribution results in a DHD of 40 %).
DHD was considered a continuous outcome variable. Data were summarized in median, range, and inter-quartile range (IQR) values (non-normally distributed). Differences in DHD between administrations with both catheter types were tested by means of a Wilcoxon signed-rank test for paired continuous data. Sample size calculation based on exploratory statistics after experiments 1–5 demonstrated that two additional experiments should be sufficient to demonstrate a statistically significant difference in DHD, with a power of 0.90 at an alpha-level of 0.05. A p-value < 0.05 was considered statistically significant. All statistics were performed in R version 3.1.1 for Mac OS X.
Fluorescent microsphere administration demonstrates catheter-dependent flow pattern
Injection force affects downstream branch targeting with a standard end-hole microcatheter
Downstream branch targeting is more consistent and homogenous with an anti-reflux catheter
Off-centered SMC position?
Deviation from homogeneous distribution
Yes (right side)
Yes (right side)
Yes (right side)
Yes (left side)
Yes (left side)
In this report, we present the first evidence that a SMC and ARC differ substantially with regard to cross-sectional catheter position, particle outflow pattern, and downstream branch distribution during the administration of microspheres in an in vitro hepatic arterial model.
The use of a SMC has long been standard clinical practice for radioembolization procedures. This may be explained by its convenient use, and the fact that potential problems with random cross-sectional catheter position or inadequate downstream branch targeting are not easily visualized during angiographic procedures. In our study, administrations with an SMC exhibited a laminar particle flow pattern. Under normal conditions, arterial blood flow is laminar. This means that blood flows in streamlines parallel to the vessel wall, without lateral mixing . The microspheres seemed to drift along these ordered streamlines. The downstream distribution therefore becomes dependent on random cross-sectional catheter positioning. The SMC induced a homogenous downstream branch distribution, only in the experiment in which it was reasonably centered. In the other 5/7 experiments, the SMC was off-centered, and a very heterogeneous downstream branch distribution was observed, with one of the targeted main branches receiving only 0-10 % of injected microspheres. Interestingly, qualitative video analysis of the first experiment also suggests that the downstream branch distribution may be dependent on injection force. We theorize that this is related to the difference between particle release velocity and blood flow velocity. Low force injection results in particle release velocities lower or equal to the local blood flow velocity. Therefore, microspheres follow the streamline into which they are released [17, 23–25]. On the contrary, high injection force results in a particle release velocity exceeding the local blood flow velocity, which allows microspheres to cross streamlines laterally.
In radioembolization treatment, it is important to achieve adequate coverage of the entire tumor-bearing liver tissue [26, 27]; therefore, a homogenous distribution over first-order branches is a minimum requirement. If the tumors are confined to a specific part of the liver, a more distal catheter position is often chosen , limiting the influence of flow on microsphere distribution. Furthermore, particle distribution is generally assumed to be consistent over repeated administrations with the catheter placed at the same position in the longitudinal axis of the vessel. In current practice, the administration of 90Y microspheres is simulated in the week(s) before treatment by injection of technetium-99 m macroaggregated albumin (99m-Tc-MAA) particles from the same longitudinal catheter position, to rule out extrahepatic microsphere deposition or significant liver-to-lung shunting . Various studies demonstrated that the 99m-Tc-MAA-distribution does not accurately predict the post-treatment intrahepatic microsphere distribution [30–32]. Inconsistencies are often ascribed to marked differences in particle characteristics such as size, density, and embolic effect , but the results of our study suggest that random differences of the cross-sectional SMC position between procedures may also play an important role.
The ARC administrations revealed a turbulent particle flow pattern. The expandable tip likely breaks up the laminar columns in the antegrade flow; and once the flow has passed the tip, flow swirls (so called eddy currents) to create a chaotic, turbulent flow pattern into which microspheres are released. Combined with the fixed centro-luminal position of the catheter orifice, this leads to a more predictable and more homogenous microsphere distribution. This may have important implications for radioembolization.
For one, tumor targeting may be improved by using the ARC. The ultimate goal is to realize an adequate radioactive dose delivery to all tumors in the liver. Two studies demonstrated that inadequate treatment of at least one tumor occurs frequently in patients treated with radioembolization, which may explain some of the inconsistencies in reported tumor response rates [12, 13, 34]. Although tumor vascularization may complicate matters in practice, we expect that a more homogenous downstream branch targeting is paramount to achieve adequate tumor coverage, which should in turn translate in to improved treatment efficacy.
Furthermore, a more consistent microsphere distribution over repeated administrations may increase the predictability of the treatment effect. In this light, the combination of the self-centering ARC and holmium-166 (166Ho) microspheres is especially interesting. Holmium-166 microspheres have been developed and clinically validated as an alternative to 90Y microspheres for radioembolization, with the added value of γ-radiation based SPECT imaging, and the option to administer a scout dose of identical 166Ho microspheres before treatment instead of 99m-Tc-MAA [12, 35]. Administering identical particles during the scout dose procedure in the same longitudinal and axial catheter position as during treatment, may be the key to accurately predict the intrahepatic distribution of the therapeutic microspheres. An accurate prediction would enable tailoring of the treatment strategy towards individual patients. Current treatment activity calculations for 90Y microspheres are based on empirically determined thresholds for a maximum tolerable whole liver absorbed dose, assuming a perfectly homogenous microsphere distribution ratio, i.e. a tumor to non-tumor (T/N) ratio of 1 . This worst-case scenario approach has been adopted, because it is currently not possible to accurately determine the individual T/N ratio prior to treatment. If the scout dose distribution truly reflects the post-treatment distribution, a minimal effective tumor dose can be targeted, while respecting a maximal tolerable healthy liver dose. Consequently, patients with a favorable T/N ratio on scout dose imaging may be treated more aggressively to enhance treatment efficacy, while a more careful approach could be selected for patients with an unfavorable T/N ratio to minimize toxicity.
Clinical studies on the use of the ARC have mainly focused on its anti-reflux characteristics, which obviates the need to coil embolize gastrointestinal branches during treatment workup, enabling a simpler and less time-consuming workflow [15, 36, 37]. In another study, it has been shown that the deployment of the ARC tip decreases blood pressure in the down-stream vascular compartment, but the consequences of this phenomenon remain uncertain . Pasciak et al. recently performed the first clinical study in which down-stream particle distribution was compared between ARC and SMC administrations, using a within-patient controlled study design. This study suggested that the use of the ARC improves selective tumor targeting, with healthy liver uptake decreasing by 24 %-89 % and tumor uptake increasing by 33 %-90 % . These results should, however, be interpreted with caution since 99mTc-MAA was used to compare the intrahepatic particle distribution, instead of therapeutic 90Y-microspheres.
Disadvantages of the ARC include a more complex catheterization compared with the SMC, and the risk of vasospasm after deployment of the catheter tip. Therefore, interventional radiologists should familiarize themselves with the special technique that is required for its use. A vasoactive drug such as nitroglycerine can be used to prevent or remedy vasospasm .
The controlled environment of the experimental model enabled us to enhance our understanding of the interplay between catheter design/positioning, injection force, and fluid-particle dynamics through observation and quantitative experiments. Clinical hemodynamics were replicated as closely as possible by using pulsatile flow with a flow rate, pressure profile and fluid viscosity based on previously published in-vivo measurements, in a 3D printed vascular branching model. Nevertheless, it is impossible to match the complexity of clinical reality. Clear differences include a more complex three-dimensional geometry of the hepatic arterial vasculature, the presence of multiple tumors that affect blood flow, the elasticity and responsiveness of arterial blood vessels, and the potential occurrence of blood flow stasis during administration of microspheres. Some of these factors, such as preferential tumor vascularization, may potentially reduce the catheter-related differences that were observed. However, considering the fact that evident differences were observed in a model with a relatively simple geometry that should facilitate a homogenous down-stream branch distribution, we expect other factors to exaggerate these differences in clinical practice.
We have initiated a clinical trial to validate the findings of this study in vivo, and determine the catheter-related effects on patient outcome (clinical trials.gov identifier: NCT02208804). In short, we will perform a phase II trial in 25 patients with irresectable, chemorefractory, and liver-dominant colorectal liver metastases. All patients will receive a pretreatment procedure during which a scout dose of 166Ho microspheres will be administered, followed by a therapeutic procedure on the same day. The effects of both catheter types will be compared within subjects, by randomly allocating the ARC to the administrations in the left or right hepatic artery, and using the SMC on the other side. This approach is justified, since the left and right hemi-livers are functionally independent, and treatment effects are limited to those areas. The primary endpoint is the T/N ratio of the radioactivity concentration on post-treatment SPECT. Secondary endpoints include absorbed tumor dose and healthy liver tissue dose, tumor response, the predictive value of the scout dose distribution, infusion efficiency, dose–response relationship, clinical toxicity and overall survival. The sample size is based on an expected factor 1.25-fold difference in T/N ratio (deemed clinically relevant): a median T/N ratio 2.0 for the ARC administrations versus 1.6 for the SMC administrations.
Using an ARC during microsphere administrations in a surrogate hepatic arterial model was associated with a favorable particle outflow pattern, a fixed centro-luminal catheter position, and a significantly more homogeneous downstream branch distribution, compared with the use of a SMC. These effects may have important implications for liver tumor treatments with radioembolization, which will be subject of investigation in a within-patient randomized controlled trial.
We thank Surefire Medical Inc. and Quirem Medical Inc. for supplying catheters and microspheres free of charge for our experiments. We also thank Chris van Kesteren for his assistance in image editing.
The Röntgen Stichting Utrecht provided an expense allowance for this study. Surefire Medical Inc. and Quirem Medical Inc. provided catheters and holmium microspheres free of charge for our experiments.
- El-Serag HB. Hepatocellular carcinoma. N Engl J Med. 2011;365:1118–27.
- Groot Koerkamp B, Fong Y. Outcomes in biliary malignancy. J Surg Oncol. 2014;110:585–91.PubMedView ArticleGoogle Scholar
- Siriwardena AK, Mason JM, Mullamitha S, Hancock HC, Jegatheeswaran S. Management of colorectal cancer presenting with synchronous liver metastases. Nat Rev Clin Oncol. 2014;11:446–59.PubMedView ArticleGoogle Scholar
- Del Prete M, Fiore F, Modica R, Marotta V, Marciello F, Ramundo V, et al. Hepatic arterial embolization in patients with neuroendocrine tumors. J Exp Clin Cancer Res. 2014;33:43.PubMed CentralPubMedView ArticleGoogle Scholar
- Lau W, Ho SKW, Yu SCH, Lai ECH, Liew C, Leung TWT. Salvage Surgery Following Downstaging of Unresectable Hepatocellular Carcinoma. Ann Surg. 2004;240:299–305.PubMed CentralPubMedView ArticleGoogle Scholar
- Lise M, Pilati P, Da Pian P, Mocellin S, Nitti D, Corazzino S. Treatment options for liver metastases from colorectal cancer. J Exp Clin Cancer Res. 2003;22(4 Suppl):149–56.PubMedGoogle Scholar
- Tonini G, Imperatori M, Vincenzi B, Frezza AM, Santini D. Rechallenge therapy and treatment holiday: different strategies in management of metastatic colorectal cancer. J Exp Clin Cancer Res. 2013;32:92.PubMed CentralPubMedView ArticleGoogle Scholar
- Spagnolo F, Caltabiano G, Queirolo P. Uveal melanoma. Cancer Treat Rev. 2012;38:549–53.PubMedView ArticleGoogle Scholar
- Khodjibekova M, Szyszko T, Singh A, Tait P, Rubello D, Al-Nahhas A. Treatment of primary and secondary liver tumours with selective internal radiation therapy. J Exp Clin Cancer Res. 2007;26:561–70.PubMedGoogle Scholar
- Lewandowski RJ, Geschwind J-F, Liapi E, Salem R. Transcatheter intraarterial therapies: rationale and overview. Radiology. 2011;259:641–57.PubMed CentralPubMedView ArticleGoogle Scholar
- Kennedy A. Radioembolization of hepatic tumors. J Gastrointest Oncol. 2014;5:178–89.PubMed CentralPubMedGoogle Scholar
- Smits MLJ, Elschot M, van den Bosch M a a J, van de Maat GH, van het Schip AD, Zonnenberg B a, Seevinck PR, Verkooijen HM, Bakker CJ, de Jong HW a M, Lam MGEH, Nijsen JFW: In vivo dosimetry based on SPECT and MR imaging of 166Ho-microspheres for treatment of liver malignancies. J Nucl Med 2013, 54:2093–100.
- Flamen P, Vanderlinden B, Delatte P, Ghanem G, Ameye L, Van Den Eynde M, et al. Multimodality imaging can predict the metabolic response of unresectable colorectal liver metastases to radioembolization therapy with Yttrium-90 labeled resin microspheres. Phys Med Biol. 2008;53:6591–603.PubMedView ArticleGoogle Scholar
- Lam MGEH, Louie JD, Abdelmaksoud MHK, Fisher G a, Cho-Phan CD, Sze DY: Limitations of body surface area-based activity calculation for radioembolization of hepatic metastases in colorectal cancer. J Vasc Interv Radiol 2014, 25:1085–93.
- Van den Hoven AF, Prince JF, Samim M, Arepally A, Zonneberg B a, Lam MGEH, van den Bosch M a a J: Posttreatment PET-CT-confirmed intrahepatic radioembolization performed without coil embolization, by using the antireflux Surefire Infusion System. Cardiovasc Intervent Radiol 2014, 37:523–8.
- Jernigan SR, Osborne J a., Mirek CJ, Buckner G: Selective Internal Radiation Therapy: Quantifying Distal Penetration and Distribution of Resin and Glass Microspheres in a Surrogate Arterial Model. J Vasc Interv Radiol 2015:1–10.
- Basciano C a, Kleinstreuer C, Kennedy AS, Dezarn W a, Childress E: Computer modeling of controlled microsphere release and targeting in a representative hepatic artery system. Ann Biomed Eng 2010, 38:1862–79.
- Hübner GH, Steudel N, Kleber G, Behrmann C, Lotterer E, Fleig WE. Hepatic arterial blood flow velocities: assessment by transcutaneous and intravascular Doppler sonography. J Hepatol. 2000;32:893–9.PubMedView ArticleGoogle Scholar
- Hirata M, Akbar SMF, Horiike N, Onji M. Noninvasive diagnosis of the degree of hepatic fibrosis using ultrasonography in patients with chronic liver disease due to hepatitis C virus. Eur J Clin Invest. 2001;31:528–35.PubMedView ArticleGoogle Scholar
- Han SB, Rice S, Cohen SM, Reynolds TB, Fong T. Duplex Doppler ultrasound of the hepatic artery in patients with acute alcoholic hepatitis. J Clin Gastroenterol. 2002;34:573–7.PubMedView ArticleGoogle Scholar
- Lowe GD, Fowkes FG, Dawes J, Donnan PT, Lennie SE, Housley E. Blood viscosity, fibrinogen, and activation of coagulation and leukocytes in peripheral arterial disease and the normal population in the Edinburgh Artery Study. Circulation. 1993;87:1915–20.PubMedView ArticleGoogle Scholar
- Ku DN. Blood Flow in Arteries. Annu Rev Fluid Mech. 1997;29:399–434.View ArticleGoogle Scholar
- Kennedy AS, Kleinstreuer C, Basciano C a, Dezarn W a. Computer modeling of yttrium-90-microsphere transport in the hepatic arterial tree to improve clinical outcomes. Int J Radiat Oncol Biol Phys. 2010;76:631–7.PubMedView ArticleGoogle Scholar
- Basciano CA, Kleinstreuer C, Kennedy AS. Computational Fluid Dynamics Modeling of 90Y Microspheres in Human Hepatic Tumors. J Nucl Med Radiat Ther. 2011;01:2–7.View ArticleGoogle Scholar
- Richards AL, Kleinstreuer C, Kennedy AS, Childress E, Buckner GD. Experimental microsphere targeting in a representative hepatic artery system. IEEE Trans Biomed Eng. 2012;59:198–204.PubMedView ArticleGoogle Scholar
- Van den Hoven AF, Smits MLJ, de Keizer B, van Leeuwen MS, van den Bosch M a a J, Lam MGEH: Identifying Aberrant Hepatic Arteries Prior to Intra-arterial Radioembolization. Cardiovasc Intervent Radiol 2014.
- Salem R, Lewandowski RJ, Sato KT, Atassi B, Ryu RK, Ibrahim S, Nemcek A a, Omary R a, Madoff DC, Murthy R: Technical aspects of radioembolization with 90Y microspheres. Tech Vasc Interv Radiol 2007, 10:12–29.
- Riaz A, Gates VL, Atassi B, Lewandowski RJ, Mulcahy MF, Ryu RK, et al. Radiation segmentectomy: a novel approach to increase safety and efficacy of radioembolization. Int J Radiat Oncol Biol Phys. 2011;79:163–71.PubMedView ArticleGoogle Scholar
- Ahmadzadehfar H, Sabet A, Biermann K, Muckle M, Brockmann H, Kuhl C, et al. The significance of 99mTc-MAA SPECT/CT liver perfusion imaging in treatment planning for 90Y-microsphere selective internal radiation treatment. J Nucl Med. 2010;51:1206–12.PubMedView ArticleGoogle Scholar
- Wondergem M, Smits MLJ, Elschot M, de Jong HW a M, Verkooijen HM, van den Bosch M a a J, Nijsen JFW, Lam MGEH: 99mTc-macroaggregated albumin poorly predicts the intrahepatic distribution of 90Y resin microspheres in hepatic radioembolization. J Nucl Med 2013, 54:1294–301.
- Dhabuwala A, Lamerton P, Stubbs RS. Relationship of 99mtechnetium labelled macroaggregated albumin (99mTc-MAA) uptake by colorectal liver metastases to response following Selective Internal Radiation Therapy (SIRT). BMC Nucl Med. 2005;5:7.PubMed CentralPubMedView ArticleGoogle Scholar
- Ulrich G, Dudeck O, Furth C, Ruf J, Grosser OS, Adolf D, et al. Predictive value of intratumoral 99mTc-macroaggregated albumin uptake in patients with colorectal liver metastases scheduled for radioembolization with 90Y-microspheres. J Nucl Med. 2013;54:516–22.PubMedView ArticleGoogle Scholar
- Van de Wiele C, Maes A, Brugman E, D’Asseler Y, De Spiegeleer B, Mees G, et al. SIRT of liver metastases: physiological and pathophysiological considerations. Eur J Nucl Med Mol Imaging. 2012;39:1646–55.PubMedView ArticleGoogle Scholar
- Van den Hoven AF, Smits MLJ, Rosenbaum CENM, Verkooijen HM, van den Bosch M a a J, Lam MGEH: The effect of intra-arterial angiotensin II on the hepatic tumor to non-tumor blood flow ratio for radioembolization: a systematic review. PLoS One 2014, 9:e86394.
- Smits MLJ, Nijsen JFW, van den Bosch M a a J, Lam MGEH, Vente M a D, Mali WPTM, van Het Schip AD, Zonnenberg B a: Holmium-166 radioembolisation in patients with unresectable, chemorefractory liver metastases (HEPAR trial): a phase 1, dose-escalation study.Lancet Oncol 2012;13:1025–34.
- Morshedi MM, Bauman M, Rose SC, Kikolski SG: Yttrium-90 Resin Microsphere Radioembolization Using an Antireflux Catheter: An Alternative to Traditional Coil Embolization for Nontarget Protection. CardiovascIntervent Radiol 2015;38(2):381–8. doi:10.1007/s00270-014-0941-y
- Fischman AM, Ward TJ, Patel RS, Arepally A, Kim E, Nowakowski FS, Lookstein RA: Prospective, Randomized Study of Coil Embolization versus Surefire Infusion System during Yttrium-90 Radioembolization with Resin Microspheres. J Vasc Interv Radiol 2014;25(11):1709–16. doi:10.1016/j.jvir.2014.08.007
- Rose SC, Kikolski SG, Chomas JE. Downstream hepatic arterial blood pressure changes caused by deployment of the surefire antireflux expandable tip. Cardiovasc Intervent Radiol. 2013;36:1262–9.PubMedView ArticleGoogle Scholar
- Pasciak AS, Mcelmurray JH, Bourgeois AC, Heidel RE, Bradley YC. The Impact of an Antireflux Catheter on Target Volume Particulate Distribution in Liver-Directed Embolotherapy : A Pilot Study. J Vasc Interv Radiol. 2015;1–10.
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.